-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170104_013051_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.819262 to 30.609160
-each separately from 7s to 30s and from 285315us to 819551us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.465764 in 1 packets - #/No: 18/18
      1.323267 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-04T01:31:01.819.262.000 to 2017-01-04T01:31:24.609.160.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.465764 in 1 packets - #/No: 18/18
      1.323267 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9319513 to 31.9318779
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.9999298 in 1 packets - #/No: 18/18
      0.9999968 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170104_013051_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.7512122 to 9.7515012
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.4656950  -- OK for SL --
   MIU time from 2017-01-04T01:31:01.819.262.000 to 2017-01-04T01:31:01.819.551.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.465764

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.2171962 to 31.2174852
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.3232630  -- OK for SL --
   MIU time from 2017-01-04T01:31:23.285.315.000 to 2017-01-04T01:31:23.285.604.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.323267

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.5407482 to 32.5410372
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-04T01:31:24.608.871.000 to 2017-01-04T01:31:24.609.160.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-04T01:31:01.819.121.750
   Max SL corrected packet time: 2017-01-04T01:31:24.609.019.750
   Min SL corrected XO rel time: 7.8192635
   Max SL corrected XO rel time: 30.6091612
   TM values from -33 to 23 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-04T01:31:01.819.121.750 (XO: 7.8192635, t=35.6°C)  Rank: 00006
   TM values: Max =    22  Min =   -30  Ave =    -4.52  Std =     7.01
   Min TM value:-30 at an offset of 100.013 usec -> 2017-01-04T01:31:01.819.221.762 (XO: 7.8193635)
   Max TM value:22 at an offset of 101.438 usec -> 2017-01-04T01:31:01.819.223.187 (XO: 7.8193650)
   Max dTM/dt [1/us]::4.24e+02 at an offset of 99.225 usec -> 2017-01-04T01:31:01.819.220.975 (XO: 7.8193627)
   Min dTM/dt [1/us]::-3.65e+02 at an offset of 173.313 usec -> 2017-01-04T01:31:01.819.295.063 (XO: 7.8194368)

 - Block number: 2 starting at 2017-01-04T01:31:23.285.174.750 (XO: 29.2853160, t=35.6°C)  Rank: 00008
   TM values: Max =    22  Min =   -33  Ave =    -4.73  Std =     9.92
   Min TM value:-33 at an offset of 137.263 usec -> 2017-01-04T01:31:23.285.312.012 (XO: 29.2854533)
   Max TM value:22 at an offset of 123.600 usec -> 2017-01-04T01:31:23.285.298.350 (XO: 29.2854396)
   Max dTM/dt [1/us]::4.40e+02 at an offset of 204.926 usec -> 2017-01-04T01:31:23.285.379.675 (XO: 29.2855209)
   Min dTM/dt [1/us]::-4.25e+02 at an offset of 122.275 usec -> 2017-01-04T01:31:23.285.297.025 (XO: 29.2854383)

 - Block number: 3 starting at 2017-01-04T01:31:24.608.730.750 (XO: 30.6088722, t=35.6°C)  Rank: 00008
   TM values: Max =    23  Min =   -33  Ave =    -4.59  Std =     8.96
   Min TM value:-33 at an offset of 190.126 usec -> 2017-01-04T01:31:24.608.920.875 (XO: 30.6090624)
   Max TM value:23 at an offset of 118.088 usec -> 2017-01-04T01:31:24.608.848.837 (XO: 30.6089903)
   Max dTM/dt [1/us]::4.33e+02 at an offset of 190.163 usec -> 2017-01-04T01:31:24.608.920.913 (XO: 30.6090624)
   Min dTM/dt [1/us]::-4.02e+02 at an offset of 46.288 usec -> 2017-01-04T01:31:24.608.777.037 (XO: 30.6089185)
