-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170103_082132_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.110191 to 35.118639
-each separately from 5s to 35s and from 110191us to 624596us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.513827 in 1 packets - #/No: 18/18
      13.493754 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-03T08:21:40.110.191.000 to 2017-01-03T08:22:10.118.639.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.513827 in 1 packets - #/No: 18/18
      13.493754 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4246511 to 36.4245554
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999489 in 1 packets - #/No: 18/18
      13.9999554 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170103_082132_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.5348426 to 6.5351316
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5137750  -- OK for SL --
   MIU time from 2017-01-03T08:21:40.110.191.000 to 2017-01-03T08:21:40.110.480.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.513827

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.0489066 to 23.0491956
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4937110  -- OK for SL --
   MIU time from 2017-01-03T08:21:56.624.307.000 to 2017-01-03T08:21:56.624.596.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.493754

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.5429066 to 36.5431956
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-03T08:22:10.118.350.000 to 2017-01-03T08:22:10.118.639.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-03T08:21:40.110.050.750
   Max SL corrected packet time: 2017-01-03T08:22:10.118.498.750
   Min SL corrected XO rel time: 5.1101919
   Max SL corrected XO rel time: 35.1186406
   TM values from -37 to 28 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-03T08:21:40.110.050.750 (XO: 5.1101919, t=35.6°C)  Rank: 00009
   TM values: Max =    25  Min =   -32  Ave =    -4.57  Std =     9.17
   Min TM value:-32 at an offset of 199.151 usec -> 2017-01-03T08:21:40.110.249.900 (XO: 5.1103910)
   Max TM value:25 at an offset of 166.538 usec -> 2017-01-03T08:21:40.110.217.288 (XO: 5.1103584)
   Max dTM/dt [1/us]::2.26e+02 at an offset of 141.863 usec -> 2017-01-03T08:21:40.110.192.612 (XO: 5.1103337)
   Min dTM/dt [1/us]::-2.36e+02 at an offset of 171.151 usec -> 2017-01-03T08:21:40.110.221.900 (XO: 5.1103630)

 - Block number: 2 starting at 2017-01-03T08:21:56.624.166.750 (XO: 21.6243086, t=35.6°C)  Rank: 00013
   TM values: Max =    28  Min =   -37  Ave =    -4.64  Std =     9.66
   Min TM value:-37 at an offset of 137.225 usec -> 2017-01-03T08:21:56.624.303.975 (XO: 21.6244458)
   Max TM value:28 at an offset of 130.488 usec -> 2017-01-03T08:21:56.624.297.237 (XO: 21.6244391)
   Max dTM/dt [1/us]::2.58e+02 at an offset of 132.988 usec -> 2017-01-03T08:21:56.624.299.737 (XO: 21.6244416)
   Min dTM/dt [1/us]::-2.62e+02 at an offset of 141.075 usec -> 2017-01-03T08:21:56.624.307.825 (XO: 21.6244497)

 - Block number: 3 starting at 2017-01-03T08:22:10.118.209.750 (XO: 35.1183516, t=35.6°C)  Rank: 00008
   TM values: Max =    23  Min =   -34  Ave =    -4.53  Std =     7.65
   Min TM value:-34 at an offset of 120.188 usec -> 2017-01-03T08:22:10.118.329.937 (XO: 35.1184718)
   Max TM value:23 at an offset of 125.563 usec -> 2017-01-03T08:22:10.118.335.312 (XO: 35.1184771)
   Max dTM/dt [1/us]::2.15e+02 at an offset of 126.825 usec -> 2017-01-03T08:22:10.118.336.575 (XO: 35.1184784)
   Min dTM/dt [1/us]::-2.06e+02 at an offset of 129.050 usec -> 2017-01-03T08:22:10.118.338.800 (XO: 35.1184806)
