-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161222_132352_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.375265 to 34.436748
-each separately from 0s to 34s and from 375265us to 573956us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.198113 in 1 packets - #/No: 18/18
      12.862503 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-22T13:23:55.375.265.000 to 2016-12-22T13:24:29.436.748.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.198113 in 1 packets - #/No: 18/18
      12.862503 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4240429 to 35.4239367
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999344 in 1 packets - #/No: 18/18
      12.9999594 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161222_132352_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7993081 to 1.7995971
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.1980470  -- OK for SL --
   MIU time from 2016-12-22T13:23:55.375.265.000 to 2016-12-22T13:23:55.375.554.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.198113

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.9976441 to 22.9979331
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8624630  -- OK for SL --
   MIU time from 2016-12-22T13:24:16.573.667.000 to 2016-12-22T13:24:16.573.956.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.862503

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.8603961 to 35.8606851
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-22T13:24:29.436.459.000 to 2016-12-22T13:24:29.436.748.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-22T13:23:55.375.124.750
   Max SL corrected packet time: 2016-12-22T13:24:29.436.607.750
   Min SL corrected XO rel time: 0.3752664
   Max SL corrected XO rel time: 34.4367498
   TM values from -36 to 29 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-22T13:23:55.375.124.750 (XO: 0.3752664, t=35.6°C)  Rank: 00008
   TM values: Max =    26  Min =   -35  Ave =    -4.48  Std =    11.66
   Min TM value:-35 at an offset of 131.988 usec -> 2016-12-22T13:23:55.375.256.737 (XO: 0.3753984)
   Max TM value:26 at an offset of 151.238 usec -> 2016-12-22T13:23:55.375.275.987 (XO: 0.3754176)
   Max dTM/dt [1/us]::2.47e+02 at an offset of 155.288 usec -> 2016-12-22T13:23:55.375.280.037 (XO: 0.3754217)
   Min dTM/dt [1/us]::-2.32e+02 at an offset of 186.788 usec -> 2016-12-22T13:23:55.375.311.538 (XO: 0.3754532)

 - Block number: 2 starting at 2016-12-22T13:24:16.573.526.750 (XO: 21.5736686, t=35.6°C)  Rank: 00009
   TM values: Max =    29  Min =   -36  Ave =    -4.52  Std =    11.61
   Min TM value:-36 at an offset of 261.101 usec -> 2016-12-22T13:24:16.573.787.850 (XO: 21.5739297)
   Max TM value:29 at an offset of 229.801 usec -> 2016-12-22T13:24:16.573.756.550 (XO: 21.5738984)
   Max dTM/dt [1/us]::2.54e+02 at an offset of 168.638 usec -> 2016-12-22T13:24:16.573.695.388 (XO: 21.5738373)
   Min dTM/dt [1/us]::-2.35e+02 at an offset of 303.463 usec -> 2016-12-22T13:24:16.573.830.213 (XO: 21.5739721)

 - Block number: 3 starting at 2016-12-22T13:24:29.436.318.750 (XO: 34.4364608, t=35.6°C)  Rank: 00006
   TM values: Max =    15  Min =   -27  Ave =    -4.50  Std =     5.79
   Min TM value:-27 at an offset of 131.113 usec -> 2016-12-22T13:24:29.436.449.862 (XO: 34.4365919)
   Max TM value:15 at an offset of 130.300 usec -> 2016-12-22T13:24:29.436.449.050 (XO: 34.4365911)
   Max dTM/dt [1/us]::1.59e+02 at an offset of 126.863 usec -> 2016-12-22T13:24:29.436.445.612 (XO: 34.4365876)
   Min dTM/dt [1/us]::-1.55e+02 at an offset of 131.000 usec -> 2016-12-22T13:24:29.436.449.750 (XO: 34.4365918)
