-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161222_115304_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.229007 to 30.790463
-each separately from 0s to 30s and from 110734us to 790463us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.881438 in 1 packets - #/No: 18/18
      15.679151 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-22T11:53:07.229.007.000 to 2016-12-22T11:53:37.790.463.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.881438 in 1 packets - #/No: 18/18
      15.679151 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4262661 to 31.4261722
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999531 in 1 packets - #/No: 18/18
      14.9999530 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161222_115304_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.6552736 to 1.6555626
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8813910  -- OK for SL --
   MIU time from 2016-12-22T11:53:07.229.007.000 to 2016-12-22T11:53:07.229.296.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.881438

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5369536 to 16.5372426
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6791030  -- OK for SL --
   MIU time from 2016-12-22T11:53:22.110.734.000 to 2016-12-22T11:53:22.111.023.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.679151

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.2163456 to 32.2166346
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-22T11:53:37.790.174.000 to 2016-12-22T11:53:37.790.463.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-22T11:53:07.228.866.750
   Max SL corrected packet time: 2016-12-22T11:53:37.790.322.750
   Min SL corrected XO rel time: 0.2290082
   Max SL corrected XO rel time: 30.7904649
   TM values from -31 to 19 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-22T11:53:07.228.866.750 (XO: 0.2290082, t=35.6°C)  Rank: 00005
   TM values: Max =    19  Min =   -31  Ave =    -4.63  Std =     7.93
   Min TM value:-31 at an offset of 133.113 usec -> 2016-12-22T11:53:07.228.999.862 (XO: 0.2291413)
   Max TM value:19 at an offset of 109.438 usec -> 2016-12-22T11:53:07.228.976.187 (XO: 0.2291176)
   Max dTM/dt [1/us]::1.99e+02 at an offset of 140.850 usec -> 2016-12-22T11:53:07.229.007.600 (XO: 0.2291491)
   Min dTM/dt [1/us]::-2.28e+02 at an offset of 0.000 usec -> 2016-12-22T11:53:07.228.866.750 (XO: 0.2290082)

 - Block number: 2 starting at 2016-12-22T11:53:22.110.593.750 (XO: 15.1107348, t=35.6°C)  Rank: 00004
   TM values: Max =    16  Min =   -26  Ave =    -4.76  Std =     6.22
   Min TM value:-26 at an offset of 77.538 usec -> 2016-12-22T11:53:22.110.671.287 (XO: 15.1108123)
   Max TM value:16 at an offset of 174.313 usec -> 2016-12-22T11:53:22.110.768.063 (XO: 15.1109091)
   Max dTM/dt [1/us]::1.65e+02 at an offset of 142.013 usec -> 2016-12-22T11:53:22.110.735.762 (XO: 15.1108768)
   Min dTM/dt [1/us]::-1.74e+02 at an offset of 142.888 usec -> 2016-12-22T11:53:22.110.736.637 (XO: 15.1108776)

 - Block number: 3 starting at 2016-12-22T11:53:37.790.033.750 (XO: 30.7901758, t=35.6°C)  Rank: 00004
   TM values: Max =    15  Min =   -25  Ave =    -4.72  Std =     5.78
   Min TM value:-25 at an offset of 161.813 usec -> 2016-12-22T11:53:37.790.195.563 (XO: 30.7903377)
   Max TM value:15 at an offset of 101.613 usec -> 2016-12-22T11:53:37.790.135.362 (XO: 30.7902775)
   Max dTM/dt [1/us]::1.63e+02 at an offset of 169.876 usec -> 2016-12-22T11:53:37.790.203.625 (XO: 30.7903457)
   Min dTM/dt [1/us]::-1.48e+02 at an offset of 222.263 usec -> 2016-12-22T11:53:37.790.256.013 (XO: 30.7903981)
