-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161219_080504_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.974101 to 42.539026
-each separately from 12s to 42s and from 538737us to 974390us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.888597 in 1 packets - #/No: 18/18
      22.675461 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-19T08:05:19.974.101.000 to 2016-12-19T08:05:49.539.026.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.888597 in 1 packets - #/No: 18/18
      22.675461 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4281534 to 43.4280603
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999783 in 1 packets - #/No: 18/18
      22.9999286 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161219_080504_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.4022536 to 14.4025426
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.8885750  -- OK for SL --
   MIU time from 2016-12-19T08:05:19.974.101.000 to 2016-12-19T08:05:19.974.390.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.888597

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.2911176 to 21.2914066
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6753910  -- OK for SL --
   MIU time from 2016-12-19T08:05:26.862.987.000 to 2016-12-19T08:05:26.863.276.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.675461

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.9667976 to 43.9670866
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-19T08:05:49.538.737.000 to 2016-12-19T08:05:49.539.026.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-19T08:05:19.973.960.750
   Max SL corrected packet time: 2016-12-19T08:05:49.538.885.750
   Min SL corrected XO rel time: 12.9741032
   Max SL corrected XO rel time: 42.5390280
   TM values from -37 to 28 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-19T08:05:19.973.960.750 (XO: 12.9741032, t=35.6°C)  Rank: 00007
   TM values: Max =    28  Min =   -37  Ave =    -4.51  Std =     9.82
   Min TM value:-37 at an offset of 137.763 usec -> 2016-12-19T08:05:19.974.098.512 (XO: 12.9742410)
   Max TM value:28 at an offset of 127.763 usec -> 2016-12-19T08:05:19.974.088.512 (XO: 12.9742310)
   Max dTM/dt [1/us]::2.72e+02 at an offset of 214.013 usec -> 2016-12-19T08:05:19.974.174.763 (XO: 12.9743172)
   Min dTM/dt [1/us]::-2.71e+02 at an offset of 113.200 usec -> 2016-12-19T08:05:19.974.073.950 (XO: 12.9742164)

 - Block number: 2 starting at 2016-12-19T08:05:26.862.846.750 (XO: 19.8629887, t=35.6°C)  Rank: 00009
   TM values: Max =    24  Min =   -33  Ave =    -4.58  Std =     8.58
   Min TM value:-33 at an offset of 124.525 usec -> 2016-12-19T08:05:26.862.971.275 (XO: 19.8631132)
   Max TM value:24 at an offset of 124.888 usec -> 2016-12-19T08:05:26.862.971.637 (XO: 19.8631135)
   Max dTM/dt [1/us]::2.67e+02 at an offset of 136.350 usec -> 2016-12-19T08:05:26.862.983.100 (XO: 19.8631250)
   Min dTM/dt [1/us]::-2.65e+02 at an offset of 116.138 usec -> 2016-12-19T08:05:26.862.962.887 (XO: 19.8631048)

 - Block number: 3 starting at 2016-12-19T08:05:49.538.596.750 (XO: 42.5387390, t=35.6°C)  Rank: 00007
   TM values: Max =    22  Min =   -33  Ave =    -4.53  Std =     8.00
   Min TM value:-33 at an offset of 124.475 usec -> 2016-12-19T08:05:49.538.721.225 (XO: 42.5388635)
   Max TM value:22 at an offset of 108.550 usec -> 2016-12-19T08:05:49.538.705.300 (XO: 42.5388476)
   Max dTM/dt [1/us]::2.70e+02 at an offset of 121.075 usec -> 2016-12-19T08:05:49.538.717.825 (XO: 42.5388601)
   Min dTM/dt [1/us]::-2.64e+02 at an offset of 119.750 usec -> 2016-12-19T08:05:49.538.716.500 (XO: 42.5388588)
