-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161217_204624_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.093220 to 33.417405
-each separately from 10s to 33s and from 93220us to 868262us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.774464 in 1 packets - #/No: 18/18
      17.548854 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-17T20:46:37.093.220.000 to 2016-12-17T20:47:00.417.405.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.774464 in 1 packets - #/No: 18/18
      17.548854 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4274305 to 34.4273598
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.9999846 in 1 packets - #/No: 18/18
      17.9999447 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      04 -> 49.8°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161217_204624_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      14 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.5206519 to 11.5209409
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.7744470  -- OK for SL --
   MIU time from 2016-12-17T20:46:37.093.220.000 to 2016-12-17T20:46:37.093.509.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.774464

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.2953879 to 17.2956769
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5487990  -- OK for SL --
   MIU time from 2016-12-17T20:46:42.867.973.000 to 2016-12-17T20:46:42.868.262.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.548854

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.8444759 to 34.8447649
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-17T20:47:00.417.116.000 to 2016-12-17T20:47:00.417.405.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-17T20:46:37.093.079.750
   Max SL corrected packet time: 2016-12-17T20:47:00.417.264.750
   Min SL corrected XO rel time: 10.0932217
   Max SL corrected XO rel time: 33.4174064
   TM values from -53 to 41 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-17T20:46:37.093.079.750 (XO: 10.0932217, t=35.6°C)  Rank: 00014
   TM values: Max =    40  Min =   -53  Ave =    -4.54  Std =    12.96
   Min TM value:-53 at an offset of 134.463 usec -> 2016-12-17T20:46:37.093.214.212 (XO: 10.0933561)
   Max TM value:40 at an offset of 132.063 usec -> 2016-12-17T20:46:37.093.211.812 (XO: 10.0933537)
   Max dTM/dt [1/us]::8.34e+02 at an offset of 134.500 usec -> 2016-12-17T20:46:37.093.214.250 (XO: 10.0933562)
   Min dTM/dt [1/us]::-7.97e+02 at an offset of 134.438 usec -> 2016-12-17T20:46:37.093.214.187 (XO: 10.0933561)

 - Block number: 2 starting at 2016-12-17T20:46:42.867.832.750 (XO: 15.8679754, t=35.6°C)  Rank: 00009
   TM values: Max =    41  Min =   -48  Ave =    -4.57  Std =    12.25
   Min TM value:-48 at an offset of 88.613 usec -> 2016-12-17T20:46:42.867.921.362 (XO: 15.8680640)
   Max TM value:41 at an offset of 121.163 usec -> 2016-12-17T20:46:42.867.953.912 (XO: 15.8680966)
   Max dTM/dt [1/us]::7.44e+02 at an offset of 107.713 usec -> 2016-12-17T20:46:42.867.940.462 (XO: 15.8680831)
   Min dTM/dt [1/us]::-6.75e+02 at an offset of 150.775 usec -> 2016-12-17T20:46:42.867.983.525 (XO: 15.8681262)

 - Block number: 3 starting at 2016-12-17T20:47:00.416.975.750 (XO: 33.4171174, t=35.6°C)  Rank: 00008
   TM values: Max =    40  Min =   -47  Ave =    -4.57  Std =    11.61
   Min TM value:-47 at an offset of 115.350 usec -> 2016-12-17T20:47:00.417.091.100 (XO: 33.4172327)
   Max TM value:40 at an offset of 117.450 usec -> 2016-12-17T20:47:00.417.093.200 (XO: 33.4172348)
   Max dTM/dt [1/us]::6.70e+02 at an offset of 117.413 usec -> 2016-12-17T20:47:00.417.093.162 (XO: 33.4172348)
   Min dTM/dt [1/us]::-7.22e+02 at an offset of 121.663 usec -> 2016-12-17T20:47:00.417.097.412 (XO: 33.4172390)
