-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161217_043752_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.997166 to 30.552808
-each separately from 6s to 30s and from 552519us to 997455us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.667462 in 1 packets - #/No: 18/18
      5.887313 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-17T04:38:01.997.166.000 to 2016-12-17T04:38:25.552.808.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.667462 in 1 packets - #/No: 18/18
      5.887313 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4285802 to 31.4285067
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999449 in 1 packets - #/No: 18/18
      5.9999816 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161217_043752_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.4257456 to 8.4260346
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6674070  -- OK for SL --
   MIU time from 2016-12-17T04:38:01.997.166.000 to 2016-12-17T04:38:01.997.455.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.667462

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.0934416 to 26.0937306
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.8872950  -- OK for SL --
   MIU time from 2016-12-17T04:38:19.664.917.000 to 2016-12-17T04:38:19.665.206.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.887313

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9810256 to 31.9813146
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-17T04:38:25.552.519.000 to 2016-12-17T04:38:25.552.808.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-17T04:38:01.997.025.750
   Max SL corrected packet time: 2016-12-17T04:38:25.552.667.750
   Min SL corrected XO rel time: 6.9971684
   Max SL corrected XO rel time: 30.5528096
   TM values from -44 to 43 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-17T04:38:01.997.025.750 (XO: 6.9971684, t=35.6°C)  Rank: 00010
   TM values: Max =    43  Min =   -44  Ave =    -4.59  Std =     7.24
   Min TM value:-44 at an offset of 189.176 usec -> 2016-12-17T04:38:01.997.214.925 (XO: 6.9973576)
   Max TM value:43 at an offset of 176.001 usec -> 2016-12-17T04:38:01.997.201.750 (XO: 6.9973444)
   Max dTM/dt [1/us]::5.85e+02 at an offset of 175.326 usec -> 2016-12-17T04:38:01.997.201.075 (XO: 6.9973437)
   Min dTM/dt [1/us]::-5.82e+02 at an offset of 181.288 usec -> 2016-12-17T04:38:01.997.207.038 (XO: 6.9973497)

 - Block number: 2 starting at 2016-12-17T04:38:19.664.776.750 (XO: 24.6649186, t=35.6°C)  Rank: 00003
   TM values: Max =    15  Min =   -25  Ave =    -4.59  Std =     4.90
   Min TM value:-25 at an offset of 133.650 usec -> 2016-12-17T04:38:19.664.910.400 (XO: 24.6650522)
   Max TM value:15 at an offset of 117.850 usec -> 2016-12-17T04:38:19.664.894.600 (XO: 24.6650364)
   Max dTM/dt [1/us]::2.43e+02 at an offset of 143.638 usec -> 2016-12-17T04:38:19.664.920.387 (XO: 24.6650622)
   Min dTM/dt [1/us]::-2.31e+02 at an offset of 259.376 usec -> 2016-12-17T04:38:19.665.036.125 (XO: 24.6651779)

 - Block number: 3 starting at 2016-12-17T04:38:25.552.378.750 (XO: 30.5525206, t=35.6°C)  Rank: 00003
   TM values: Max =     9  Min =   -18  Ave =    -4.61  Std =     3.57
   Min TM value:-18 at an offset of 21.625 usec -> 2016-12-17T04:38:25.552.400.375 (XO: 30.5525422)
   Max TM value:9 at an offset of 196.038 usec -> 2016-12-17T04:38:25.552.574.788 (XO: 30.5527167)
   Max dTM/dt [1/us]::1.38e+02 at an offset of 248.088 usec -> 2016-12-17T04:38:25.552.626.838 (XO: 30.5527687)
   Min dTM/dt [1/us]::-1.65e+02 at an offset of 240.563 usec -> 2016-12-17T04:38:25.552.619.313 (XO: 30.5527612)
