-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161216_220424_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.004361 to 38.299887
-each separately from 0s to 38s and from 4361us to 954444us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.949505 in 1 packets - #/No: 18/18
      16.345154 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-16T22:04:27.004.361.000 to 2016-12-16T22:05:05.299.887.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.949505 in 1 packets - #/No: 18/18
      16.345154 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9304941 to 39.9303768
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999352 in 1 packets - #/No: 18/18
      16.9999475 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161216_220424_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9348558 to 1.9351448
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.9494390  -- OK for SL --
   MIU time from 2016-12-16T22:04:27.004.361.000 to 2016-12-16T22:04:27.004.650.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.949505

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      19 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.8845838 to 23.8848728
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3451030  -- OK for SL --
   MIU time from 2016-12-16T22:04:48.954.155.000 to 2016-12-16T22:04:48.954.444.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.345154

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.2299758 to 40.2302648
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-16T22:05:05.299.598.000 to 2016-12-16T22:05:05.299.887.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-16T22:04:27.004.220.750
   Max SL corrected packet time: 2016-12-16T22:05:05.299.746.750
   Min SL corrected XO rel time: 0.0043617
   Max SL corrected XO rel time: 38.2998889
   TM values from -50 to 34 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-16T22:04:27.004.220.750 (XO: 0.0043617, t=37.2°C)  Rank: 00009
   TM values: Max =    34  Min =   -42  Ave =    -4.67  Std =    10.57
   Min TM value:-42 at an offset of 121.363 usec -> 2016-12-16T22:04:27.004.342.112 (XO: 0.0044831)
   Max TM value:34 at an offset of 260.613 usec -> 2016-12-16T22:04:27.004.481.363 (XO: 0.0046223)
   Max dTM/dt [1/us]::5.30e+02 at an offset of 164.750 usec -> 2016-12-16T22:04:27.004.385.500 (XO: 0.0045265)
   Min dTM/dt [1/us]::-5.23e+02 at an offset of 165.638 usec -> 2016-12-16T22:04:27.004.386.387 (XO: 0.0045274)

 - Block number: 2 starting at 2016-12-16T22:04:48.954.014.750 (XO: 21.9541575, t=37.2°C)  Rank: 00019
   TM values: Max =    30  Min =   -50  Ave =    -4.56  Std =    10.49
   Min TM value:-50 at an offset of 101.925 usec -> 2016-12-16T22:04:48.954.116.675 (XO: 21.9542594)
   Max TM value:30 at an offset of 111.825 usec -> 2016-12-16T22:04:48.954.126.575 (XO: 21.9542693)
   Max dTM/dt [1/us]::4.48e+02 at an offset of 301.838 usec -> 2016-12-16T22:04:48.954.316.588 (XO: 21.9544593)
   Min dTM/dt [1/us]::-4.19e+02 at an offset of 278.126 usec -> 2016-12-16T22:04:48.954.292.875 (XO: 21.9544356)

 - Block number: 3 starting at 2016-12-16T22:05:05.299.457.750 (XO: 38.2995999, t=37.2°C)  Rank: 00010
   TM values: Max =    34  Min =   -44  Ave =    -4.76  Std =    10.16
   Min TM value:-44 at an offset of 97.738 usec -> 2016-12-16T22:05:05.299.555.487 (XO: 38.2996977)
   Max TM value:34 at an offset of 99.988 usec -> 2016-12-16T22:05:05.299.557.737 (XO: 38.2996999)
   Max dTM/dt [1/us]::5.48e+02 at an offset of 97.775 usec -> 2016-12-16T22:05:05.299.555.525 (XO: 38.2996977)
   Min dTM/dt [1/us]::-5.27e+02 at an offset of 98.650 usec -> 2016-12-16T22:05:05.299.556.400 (XO: 38.2996986)
