-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161216_193304_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.704610 to 34.962483
-each separately from 13s to 34s and from 296775us to 962483us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      1.591876 in 1 packets - #/No: 18/18
      19.665130 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-16T19:33:20.704.610.000 to 2016-12-16T19:33:41.962.483.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      1.591876 in 1 packets - #/No: 18/18
      19.665130 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4296490 to 35.4295853
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      1.9999939 in 1 packets - #/No: 18/18
      18.9999424 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      03 -> 51.1°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C1 -> 37.2°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161216_193304_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      22 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.1342589 to 15.1345479
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.5918710  -- OK for SL --
   MIU time from 2016-12-16T19:33:20.704.610.000 to 2016-12-16T19:33:20.704.899.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.591876

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      19 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7264189 to 16.7267079
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.6650710  -- OK for SL --
   MIU time from 2016-12-16T19:33:22.296.775.000 to 2016-12-16T19:33:22.297.064.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.665130

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.3917789 to 36.3920679
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-16T19:33:41.962.194.000 to 2016-12-16T19:33:41.962.483.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-16T19:33:20.704.469.750
   Max SL corrected packet time: 2016-12-16T19:33:41.962.342.750
   Min SL corrected XO rel time: 13.7046121
   Max SL corrected XO rel time: 34.9624855
   TM values from -62 to 49 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-16T19:33:20.704.469.750 (XO: 13.7046121, t=37.2°C)  Rank: 00022
   TM values: Max =    49  Min =   -56  Ave =    -4.66  Std =    14.98
   Min TM value:-56 at an offset of 124.788 usec -> 2016-12-16T19:33:20.704.594.537 (XO: 13.7047369)
   Max TM value:49 at an offset of 134.575 usec -> 2016-12-16T19:33:20.704.604.325 (XO: 13.7047467)
   Max dTM/dt [1/us]::8.93e+02 at an offset of 134.538 usec -> 2016-12-16T19:33:20.704.604.287 (XO: 13.7047466)
   Min dTM/dt [1/us]::-9.03e+02 at an offset of 125.588 usec -> 2016-12-16T19:33:20.704.595.337 (XO: 13.7047377)

 - Block number: 2 starting at 2016-12-16T19:33:22.296.634.750 (XO: 15.2967769, t=37.2°C)  Rank: 00019
   TM values: Max =    47  Min =   -62  Ave =    -4.67  Std =    13.41
   Min TM value:-62 at an offset of 135.263 usec -> 2016-12-16T19:33:22.296.770.012 (XO: 15.2969122)
   Max TM value:47 at an offset of 91.625 usec -> 2016-12-16T19:33:22.296.726.375 (XO: 15.2968685)
   Max dTM/dt [1/us]::8.22e+02 at an offset of 136.638 usec -> 2016-12-16T19:33:22.296.771.387 (XO: 15.2969135)
   Min dTM/dt [1/us]::-8.16e+02 at an offset of 135.225 usec -> 2016-12-16T19:33:22.296.769.975 (XO: 15.2969121)

 - Block number: 3 starting at 2016-12-16T19:33:41.962.053.750 (XO: 34.9621965, t=37.2°C)  Rank: 00011
   TM values: Max =    38  Min =   -49  Ave =    -4.52  Std =    10.57
   Min TM value:-49 at an offset of 129.413 usec -> 2016-12-16T19:33:41.962.183.162 (XO: 34.9623259)
   Max TM value:38 at an offset of 128.838 usec -> 2016-12-16T19:33:41.962.182.587 (XO: 34.9623253)
   Max dTM/dt [1/us]::5.87e+02 at an offset of 126.625 usec -> 2016-12-16T19:33:41.962.180.375 (XO: 34.9623231)
   Min dTM/dt [1/us]::-5.55e+02 at an offset of 127.700 usec -> 2016-12-16T19:33:41.962.181.450 (XO: 34.9623242)
