-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161216_140008_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.373448 to 30.030087
-each separately from 10s to 30s and from 29798us to 373737us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.832751 in 1 packets - #/No: 18/18
      14.823021 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-16T14:00:21.373.448.000 to 2016-12-16T14:00:41.030.087.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.832751 in 1 packets - #/No: 18/18
      14.823021 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4292266 to 31.4291646
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.9999845 in 1 packets - #/No: 18/18
      14.9999535 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161216_140008_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.8026753 to 11.8029643
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.8327350  -- OK for SL --
   MIU time from 2016-12-16T14:00:21.373.448.000 to 2016-12-16T14:00:21.373.737.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.832751

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6356993 to 16.6359883
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8229750  -- OK for SL --
   MIU time from 2016-12-16T14:00:26.206.488.000 to 2016-12-16T14:00:26.206.777.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.823021

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4589633 to 31.4592523
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-16T14:00:41.029.798.000 to 2016-12-16T14:00:41.030.087.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-16T14:00:21.373.307.750
   Max SL corrected packet time: 2016-12-16T14:00:41.029.946.750
   Min SL corrected XO rel time: 10.3734499
   Max SL corrected XO rel time: 30.0300878
   TM values from -29 to 19 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-16T14:00:21.373.307.750 (XO: 10.3734499, t=35.6°C)  Rank: 00007
   TM values: Max =    19  Min =   -29  Ave =    -4.72  Std =     7.93
   Min TM value:-29 at an offset of 172.663 usec -> 2016-12-16T14:00:21.373.480.413 (XO: 10.3736225)
   Max TM value:19 at an offset of 124.775 usec -> 2016-12-16T14:00:21.373.432.525 (XO: 10.3735746)
   Max dTM/dt [1/us]::2.00e+02 at an offset of 128.950 usec -> 2016-12-16T14:00:21.373.436.700 (XO: 10.3735788)
   Min dTM/dt [1/us]::-1.97e+02 at an offset of 146.413 usec -> 2016-12-16T14:00:21.373.454.162 (XO: 10.3735963)

 - Block number: 2 starting at 2016-12-16T14:00:26.206.347.750 (XO: 15.2064888, t=35.6°C)  Rank: 00004
   TM values: Max =    19  Min =   -24  Ave =    -4.69  Std =     5.47
   Min TM value:-24 at an offset of 31.538 usec -> 2016-12-16T14:00:26.206.379.287 (XO: 15.2065204)
   Max TM value:19 at an offset of 82.613 usec -> 2016-12-16T14:00:26.206.430.362 (XO: 15.2065715)
   Max dTM/dt [1/us]::1.21e+02 at an offset of 126.325 usec -> 2016-12-16T14:00:26.206.474.075 (XO: 15.2066152)
   Min dTM/dt [1/us]::-1.36e+02 at an offset of 31.488 usec -> 2016-12-16T14:00:26.206.379.237 (XO: 15.2065203)

 - Block number: 3 starting at 2016-12-16T14:00:41.029.657.750 (XO: 30.0297988, t=35.6°C)  Rank: 00004
   TM values: Max =    12  Min =   -23  Ave =    -4.59  Std =     5.17
   Min TM value:-23 at an offset of 138.588 usec -> 2016-12-16T14:00:41.029.796.337 (XO: 30.0299374)
   Max TM value:12 at an offset of 136.450 usec -> 2016-12-16T14:00:41.029.794.200 (XO: 30.0299352)
   Max dTM/dt [1/us]::1.24e+02 at an offset of 38.588 usec -> 2016-12-16T14:00:41.029.696.337 (XO: 30.0298374)
   Min dTM/dt [1/us]::-1.88e+02 at an offset of 92.825 usec -> 2016-12-16T14:00:41.029.750.575 (XO: 30.0298916)
