-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161214_063047_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.740247 to 35.869455
-each separately from 12s to 35s and from 256155us to 869455us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.515619 in 1 packets - #/No: 18/18
      11.612722 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-14T06:31:02.740.247.000 to 2016-12-14T06:31:25.869.455.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.515619 in 1 packets - #/No: 18/18
      11.612722 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9279223 to 36.9278517
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999632 in 1 packets - #/No: 18/18
      10.9999662 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161214_063047_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.6681688 to 14.6684578
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.5155830  -- OK for SL --
   MIU time from 2016-12-14T06:31:02.740.247.000 to 2016-12-14T06:31:02.740.536.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.515619

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.1840408 to 26.1843298
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.6126870  -- OK for SL --
   MIU time from 2016-12-14T06:31:14.256.155.000 to 2016-12-14T06:31:14.256.444.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.612722

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.7970168 to 37.7973058
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-14T06:31:25.869.166.000 to 2016-12-14T06:31:25.869.455.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-14T06:31:02.740.106.750
   Max SL corrected packet time: 2016-12-14T06:31:25.869.314.750
   Min SL corrected XO rel time: 12.7402487
   Max SL corrected XO rel time: 35.8694567
   TM values from -41 to 30 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-14T06:31:02.740.106.750 (XO: 12.7402487, t=35.6°C)  Rank: 00007
   TM values: Max =    24  Min =   -32  Ave =    -4.63  Std =     7.71
   Min TM value:-32 at an offset of 160.938 usec -> 2016-12-14T06:31:02.740.267.687 (XO: 12.7404097)
   Max TM value:24 at an offset of 120.563 usec -> 2016-12-14T06:31:02.740.227.312 (XO: 12.7403693)
   Max dTM/dt [1/us]::3.93e+02 at an offset of 113.863 usec -> 2016-12-14T06:31:02.740.220.612 (XO: 12.7403626)
   Min dTM/dt [1/us]::-4.11e+02 at an offset of 127.113 usec -> 2016-12-14T06:31:02.740.233.862 (XO: 12.7403758)

 - Block number: 2 starting at 2016-12-14T06:31:14.256.014.750 (XO: 24.2561561, t=35.6°C)  Rank: 00009
   TM values: Max =    27  Min =   -35  Ave =    -4.62  Std =     9.54
   Min TM value:-35 at an offset of 113.325 usec -> 2016-12-14T06:31:14.256.128.075 (XO: 24.2562694)
   Max TM value:27 at an offset of 130.800 usec -> 2016-12-14T06:31:14.256.145.550 (XO: 24.2562869)
   Max dTM/dt [1/us]::4.70e+02 at an offset of 129.125 usec -> 2016-12-14T06:31:14.256.143.875 (XO: 24.2562852)
   Min dTM/dt [1/us]::-4.83e+02 at an offset of 128.238 usec -> 2016-12-14T06:31:14.256.142.987 (XO: 24.2562843)

 - Block number: 3 starting at 2016-12-14T06:31:25.869.025.750 (XO: 35.8691677, t=35.6°C)  Rank: 00007
   TM values: Max =    30  Min =   -41  Ave =    -4.53  Std =     9.56
   Min TM value:-41 at an offset of 138.413 usec -> 2016-12-14T06:31:25.869.164.162 (XO: 35.8693061)
   Max TM value:30 at an offset of 163.525 usec -> 2016-12-14T06:31:25.869.189.275 (XO: 35.8693312)
   Max dTM/dt [1/us]::5.13e+02 at an offset of 138.450 usec -> 2016-12-14T06:31:25.869.164.200 (XO: 35.8693062)
   Min dTM/dt [1/us]::-4.84e+02 at an offset of 133.338 usec -> 2016-12-14T06:31:25.869.159.087 (XO: 35.8693010)
