-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161211_165815_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.315077 to 31.247710
-each separately from 7s to 31s and from 247421us to 481630us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.165975 in 1 packets - #/No: 18/18
      15.765791 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-11T16:58:25.315.077.000 to 2016-12-11T16:58:49.247.710.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.165975 in 1 packets - #/No: 18/18
      15.765791 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4282597 to 32.4281877
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999760 in 1 packets - #/No: 18/18
      15.9999520 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161211_165815_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7433372 to 8.7436262
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.1659510  -- OK for SL --
   MIU time from 2016-12-11T16:58:25.315.077.000 to 2016-12-11T16:58:25.315.366.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.165975

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9095772 to 16.9098662
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7657430  -- OK for SL --
   MIU time from 2016-12-11T16:58:33.481.341.000 to 2016-12-11T16:58:33.481.630.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.765791

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.6756092 to 32.6758982
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-11T16:58:49.247.421.000 to 2016-12-11T16:58:49.247.710.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-11T16:58:25.314.936.750
   Max SL corrected packet time: 2016-12-11T16:58:49.247.569.750
   Min SL corrected XO rel time: 7.3150784
   Max SL corrected XO rel time: 31.2477113
   TM values from -31 to 22 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-11T16:58:25.314.936.750 (XO: 7.3150784, t=35.6°C)  Rank: 00007
   TM values: Max =    19  Min =   -27  Ave =    -4.64  Std =     6.57
   Min TM value:-27 at an offset of 130.988 usec -> 2016-12-11T16:58:25.315.067.737 (XO: 7.3152094)
   Max TM value:19 at an offset of 187.963 usec -> 2016-12-11T16:58:25.315.124.713 (XO: 7.3152664)
   Max dTM/dt [1/us]::2.19e+02 at an offset of 166.000 usec -> 2016-12-11T16:58:25.315.102.750 (XO: 7.3152444)
   Min dTM/dt [1/us]::-2.56e+02 at an offset of 145.688 usec -> 2016-12-11T16:58:25.315.082.437 (XO: 7.3152241)

 - Block number: 2 starting at 2016-12-11T16:58:33.481.200.750 (XO: 15.4813429, t=35.6°C)  Rank: 00006
   TM values: Max =    20  Min =   -30  Ave =    -4.71  Std =     7.24
   Min TM value:-30 at an offset of 128.188 usec -> 2016-12-11T16:58:33.481.328.937 (XO: 15.4814711)
   Max TM value:20 at an offset of 121.450 usec -> 2016-12-11T16:58:33.481.322.200 (XO: 15.4814644)
   Max dTM/dt [1/us]::3.15e+02 at an offset of 4.975 usec -> 2016-12-11T16:58:33.481.205.725 (XO: 15.4813479)
   Min dTM/dt [1/us]::-2.96e+02 at an offset of 274.463 usec -> 2016-12-11T16:58:33.481.475.213 (XO: 15.4816174)

 - Block number: 3 starting at 2016-12-11T16:58:49.247.280.750 (XO: 31.2474223, t=35.6°C)  Rank: 00006
   TM values: Max =    22  Min =   -31  Ave =    -4.72  Std =     7.70
   Min TM value:-31 at an offset of 140.200 usec -> 2016-12-11T16:58:49.247.420.950 (XO: 31.2475625)
   Max TM value:22 at an offset of 27.050 usec -> 2016-12-11T16:58:49.247.307.800 (XO: 31.2474493)
   Max dTM/dt [1/us]::2.69e+02 at an offset of 257.638 usec -> 2016-12-11T16:58:49.247.538.388 (XO: 31.2476799)
   Min dTM/dt [1/us]::-2.99e+02 at an offset of 89.538 usec -> 2016-12-11T16:58:49.247.370.287 (XO: 31.2475118)
