-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161211_062239_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.016870 to 34.316886
-each separately from 2s to 34s and from 16870us to 931239us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.913791 in 1 packets - #/No: 18/18
      10.385358 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-11T06:22:44.016.870.000 to 2016-12-11T06:23:16.316.886.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.913791 in 1 packets - #/No: 18/18
      10.385358 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4257974 to 35.4257020
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999374 in 1 packets - #/No: 18/18
      10.9999672 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161211_062239_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.4426681 to 3.4429571
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.9137270  -- OK for SL --
   MIU time from 2016-12-11T06:22:44.016.870.000 to 2016-12-11T06:22:44.017.159.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.913791

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.3566841 to 25.3569731
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.3853270  -- OK for SL --
   MIU time from 2016-12-11T06:23:05.930.950.000 to 2016-12-11T06:23:05.931.239.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.385358

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.7423001 to 35.7425891
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-11T06:23:16.316.597.000 to 2016-12-11T06:23:16.316.886.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-11T06:22:44.016.729.750
   Max SL corrected packet time: 2016-12-11T06:23:16.316.745.750
   Min SL corrected XO rel time: 2.0168708
   Max SL corrected XO rel time: 34.3168880
   TM values from -31 to 22 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-11T06:22:44.016.729.750 (XO: 2.0168708, t=37.2°C)  Rank: 00006
   TM values: Max =    22  Min =   -31  Ave =    -4.69  Std =     7.11
   Min TM value:-31 at an offset of 121.950 usec -> 2016-12-11T06:22:44.016.851.700 (XO: 2.0169927)
   Max TM value:22 at an offset of 159.238 usec -> 2016-12-11T06:22:44.016.888.987 (XO: 2.0170300)
   Max dTM/dt [1/us]::2.79e+02 at an offset of 113.413 usec -> 2016-12-11T06:22:44.016.843.162 (XO: 2.0169842)
   Min dTM/dt [1/us]::-2.88e+02 at an offset of 100.175 usec -> 2016-12-11T06:22:44.016.829.925 (XO: 2.0169709)

 - Block number: 2 starting at 2016-12-11T06:23:05.930.809.750 (XO: 23.9309521, t=37.2°C)  Rank: 00006
   TM values: Max =    18  Min =   -27  Ave =    -4.66  Std =     6.16
   Min TM value:-27 at an offset of 95.375 usec -> 2016-12-11T06:23:05.930.905.125 (XO: 23.9310475)
   Max TM value:18 at an offset of 103.600 usec -> 2016-12-11T06:23:05.930.913.350 (XO: 23.9310557)
   Max dTM/dt [1/us]::2.13e+02 at an offset of 117.275 usec -> 2016-12-11T06:23:05.930.927.025 (XO: 23.9310694)
   Min dTM/dt [1/us]::-2.38e+02 at an offset of 95.325 usec -> 2016-12-11T06:23:05.930.905.075 (XO: 23.9310474)

 - Block number: 3 starting at 2016-12-11T06:23:16.316.456.750 (XO: 34.3165990, t=37.2°C)  Rank: 00006
   TM values: Max =    19  Min =   -27  Ave =    -4.71  Std =     6.79
   Min TM value:-27 at an offset of 77.025 usec -> 2016-12-11T06:23:16.316.533.775 (XO: 34.3166760)
   Max TM value:19 at an offset of 120.400 usec -> 2016-12-11T06:23:16.316.577.150 (XO: 34.3167194)
   Max dTM/dt [1/us]::2.28e+02 at an offset of 235.776 usec -> 2016-12-11T06:23:16.316.692.525 (XO: 34.3168348)
   Min dTM/dt [1/us]::-2.48e+02 at an offset of 275.638 usec -> 2016-12-11T06:23:16.316.732.388 (XO: 34.3168747)
