-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161209_143135_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.958509 to 32.095195
-each separately from 0s to 32s and from 94906us to 958798us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.890158 in 1 packets - #/No: 18/18
      15.245661 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-09T14:31:38.958.509.000 to 2016-12-09T14:32:10.095.195.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.890158 in 1 packets - #/No: 18/18
      15.245661 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4267982 to 33.4267033
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999526 in 1 packets - #/No: 18/18
      15.9999525 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      04 -> 49.8°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      03 -> 51.1°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161209_143135_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3853063 to 2.3855953
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8901110  -- OK for SL --
   MIU time from 2016-12-09T14:31:38.958.509.000 to 2016-12-09T14:31:38.958.798.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.890158

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.2757063 to 18.2759953
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.2456150  -- OK for SL --
   MIU time from 2016-12-09T14:31:54.848.956.000 to 2016-12-09T14:31:54.849.245.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.245661

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.5216103 to 33.5218993
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-09T14:32:10.094.906.000 to 2016-12-09T14:32:10.095.195.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-09T14:31:38.958.368.750
   Max SL corrected packet time: 2016-12-09T14:32:10.095.054.750
   Min SL corrected XO rel time: 0.9585110
   Max SL corrected XO rel time: 32.0951963
   TM values from -35 to 26 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-09T14:31:38.958.368.750 (XO: 0.9585110, t=35.6°C)  Rank: 00005
   TM values: Max =    22  Min =   -32  Ave =    -4.59  Std =     7.67
   Min TM value:-32 at an offset of 233.051 usec -> 2016-12-09T14:31:38.958.601.800 (XO: 0.9587440)
   Max TM value:22 at an offset of 185.076 usec -> 2016-12-09T14:31:38.958.553.825 (XO: 0.9586961)
   Max dTM/dt [1/us]::1.70e+02 at an offset of 180.463 usec -> 2016-12-09T14:31:38.958.549.213 (XO: 0.9586914)
   Min dTM/dt [1/us]::-1.82e+02 at an offset of 226.338 usec -> 2016-12-09T14:31:38.958.595.088 (XO: 0.9587373)

 - Block number: 2 starting at 2016-12-09T14:31:54.848.815.750 (XO: 16.8489580, t=35.6°C)  Rank: 00005
   TM values: Max =    18  Min =   -31  Ave =    -4.64  Std =     6.66
   Min TM value:-31 at an offset of 128.425 usec -> 2016-12-09T14:31:54.848.944.175 (XO: 16.8490865)
   Max TM value:18 at an offset of 162.750 usec -> 2016-12-09T14:31:54.848.978.500 (XO: 16.8491208)
   Max dTM/dt [1/us]::2.10e+02 at an offset of 122.525 usec -> 2016-12-09T14:31:54.848.938.275 (XO: 16.8490806)
   Min dTM/dt [1/us]::-2.05e+02 at an offset of 117.238 usec -> 2016-12-09T14:31:54.848.932.987 (XO: 16.8490753)

 - Block number: 3 starting at 2016-12-09T14:32:10.094.765.750 (XO: 32.0949073, t=35.6°C)  Rank: 00005
   TM values: Max =    26  Min =   -35  Ave =    -4.49  Std =     8.53
   Min TM value:-35 at an offset of 177.713 usec -> 2016-12-09T14:32:10.094.943.463 (XO: 32.0950850)
   Max TM value:26 at an offset of 164.600 usec -> 2016-12-09T14:32:10.094.930.350 (XO: 32.0950719)
   Max dTM/dt [1/us]::2.35e+02 at an offset of 107.888 usec -> 2016-12-09T14:32:10.094.873.637 (XO: 32.0950152)
   Min dTM/dt [1/us]::-2.38e+02 at an offset of 73.563 usec -> 2016-12-09T14:32:10.094.839.312 (XO: 32.0949808)
