-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161209_075807_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.320808 to 34.241527
-each separately from 13s to 34s and from 241238us to 935223us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.613837 in 1 packets - #/No: 18/18
      10.306015 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-09T07:58:23.320.808.000 to 2016-12-09T07:58:44.241.527.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.613837 in 1 packets - #/No: 18/18
      10.306015 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4269640 to 35.4269014
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.9999702 in 1 packets - #/No: 18/18
      10.9999672 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      03 -> 51.1°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161209_075807_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.7477720 to 14.7480610
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6138070  -- OK for SL --
   MIU time from 2016-12-09T07:58:23.320.808.000 to 2016-12-09T07:58:23.321.097.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.613837

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.3618680 to 25.3621570
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.3059830  -- OK for SL --
   MIU time from 2016-12-09T07:58:33.934.934.000 to 2016-12-09T07:58:33.935.223.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.306015

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.6681400 to 35.6684290
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-09T07:58:44.241.238.000 to 2016-12-09T07:58:44.241.527.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-09T07:58:23.320.667.750
   Max SL corrected packet time: 2016-12-09T07:58:44.241.386.750
   Min SL corrected XO rel time: 13.3208090
   Max SL corrected XO rel time: 34.2415283
   TM values from -69 to 59 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-09T07:58:23.320.667.750 (XO: 13.3208090, t=35.6°C)  Rank: 00011
   TM values: Max =    59  Min =   -69  Ave =    -4.61  Std =    17.63
   Min TM value:-69 at an offset of 190.026 usec -> 2016-12-09T07:58:23.320.857.775 (XO: 13.3209990)
   Max TM value:59 at an offset of 190.776 usec -> 2016-12-09T07:58:23.320.858.525 (XO: 13.3209997)
   Max dTM/dt [1/us]::5.65e+02 at an offset of 191.251 usec -> 2016-12-09T07:58:23.320.859.000 (XO: 13.3210002)
   Min dTM/dt [1/us]::-5.53e+02 at an offset of 195.113 usec -> 2016-12-09T07:58:23.320.862.863 (XO: 13.3210041)

 - Block number: 2 starting at 2016-12-09T07:58:33.934.793.750 (XO: 23.9349366, t=35.6°C)  Rank: 00013
   TM values: Max =    51  Min =   -61  Ave =    -4.62  Std =    15.33
   Min TM value:-61 at an offset of 222.276 usec -> 2016-12-09T07:58:33.935.016.025 (XO: 23.9351589)
   Max TM value:51 at an offset of 112.100 usec -> 2016-12-09T07:58:33.934.905.850 (XO: 23.9350487)
   Max dTM/dt [1/us]::4.46e+02 at an offset of 142.788 usec -> 2016-12-09T07:58:33.934.936.537 (XO: 23.9350794)
   Min dTM/dt [1/us]::-4.65e+02 at an offset of 107.800 usec -> 2016-12-09T07:58:33.934.901.550 (XO: 23.9350444)

 - Block number: 3 starting at 2016-12-09T07:58:44.241.097.750 (XO: 34.2412393, t=35.6°C)  Rank: 00012
   TM values: Max =    42  Min =   -56  Ave =    -4.68  Std =    14.04
   Min TM value:-56 at an offset of 244.963 usec -> 2016-12-09T07:58:44.241.342.713 (XO: 34.2414843)
   Max TM value:42 at an offset of 51.250 usec -> 2016-12-09T07:58:44.241.149.000 (XO: 34.2412906)
   Max dTM/dt [1/us]::4.87e+02 at an offset of 241.563 usec -> 2016-12-09T07:58:44.241.339.313 (XO: 34.2414809)
   Min dTM/dt [1/us]::-5.10e+02 at an offset of 45.763 usec -> 2016-12-09T07:58:44.241.143.512 (XO: 34.2412851)
