-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20161207_203927_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.190953 to 31.643518
-each separately from 3s to 31s and from 190953us to 922564us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.731033 in 1 packets - #/No: 18/18
      8.720665 in 1 packets - #/No: 36/18
MIU packet time from 2016-12-07T20:39:33.190.953.000 to 2016-12-07T20:40:01.643.518.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.731033 in 1 packets - #/No: 18/18
      8.720665 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4278141 to 32.4277317
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.9999440 in 1 packets - #/No: 18/18
      8.9999736 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C1 -> 37.2°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20161207_203927_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      17 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.6187682 to 4.6190572
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.7309750  -- OK for SL --
   MIU time from 2016-12-07T20:39:33.190.953.000 to 2016-12-07T20:39:33.191.242.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.731033

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      16 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.3500322 to 24.3503212
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7206390  -- OK for SL --
   MIU time from 2016-12-07T20:39:52.922.275.000 to 2016-12-07T20:39:52.922.564.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.720665

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.0709602 to 33.0712492
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2016-12-07T20:40:01.643.229.000 to 2016-12-07T20:40:01.643.518.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2016-12-07T20:39:33.190.812.750
   Max SL corrected packet time: 2016-12-07T20:40:01.643.377.750
   Min SL corrected XO rel time: 3.1909547
   Max SL corrected XO rel time: 31.6435194
   TM values from -75 to 59 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2016-12-07T20:39:33.190.812.750 (XO: 3.1909547, t=37.2°C)  Rank: 00017
   TM values: Max =    59  Min =   -75  Ave =    -4.69  Std =    14.56
   Min TM value:-75 at an offset of 191.388 usec -> 2016-12-07T20:39:33.191.004.138 (XO: 3.1911461)
   Max TM value:59 at an offset of 192.276 usec -> 2016-12-07T20:39:33.191.005.025 (XO: 3.1911469)
   Max dTM/dt [1/us]::1.02e+03 at an offset of 192.101 usec -> 2016-12-07T20:39:33.191.004.850 (XO: 3.1911468)
   Min dTM/dt [1/us]::-1.01e+03 at an offset of 191.351 usec -> 2016-12-07T20:39:33.191.004.100 (XO: 3.1911460)

 - Block number: 2 starting at 2016-12-07T20:39:52.922.134.750 (XO: 22.9222768, t=37.2°C)  Rank: 00016
   TM values: Max =    46  Min =   -50  Ave =    -4.74  Std =    13.46
   Min TM value:-50 at an offset of 114.575 usec -> 2016-12-07T20:39:52.922.249.325 (XO: 22.9223913)
   Max TM value:46 at an offset of 244.413 usec -> 2016-12-07T20:39:52.922.379.163 (XO: 22.9225212)
   Max dTM/dt [1/us]::7.35e+02 at an offset of 43.563 usec -> 2016-12-07T20:39:52.922.178.312 (XO: 22.9223203)
   Min dTM/dt [1/us]::-6.96e+02 at an offset of 120.100 usec -> 2016-12-07T20:39:52.922.254.850 (XO: 22.9223969)

 - Block number: 3 starting at 2016-12-07T20:40:01.643.088.750 (XO: 31.6432304, t=37.2°C)  Rank: 00013
   TM values: Max =    42  Min =   -54  Ave =    -4.75  Std =    13.60
   Min TM value:-54 at an offset of 192.438 usec -> 2016-12-07T20:40:01.643.281.188 (XO: 31.6434229)
   Max TM value:42 at an offset of 107.788 usec -> 2016-12-07T20:40:01.643.196.537 (XO: 31.6433382)
   Max dTM/dt [1/us]::7.68e+02 at an offset of 120.875 usec -> 2016-12-07T20:40:01.643.209.625 (XO: 31.6433513)
   Min dTM/dt [1/us]::-7.42e+02 at an offset of 120.950 usec -> 2016-12-07T20:40:01.643.209.700 (XO: 31.6433514)
