-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151203_120846_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.018229 to 30.026004
-each separately from 0s to 30s and from 11844us to 26004us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.993326 in 1 packets - #/No: 18/18
      15.013582 in 1 packets - #/No: 36/18
MIU packet time from 2015-12-03T12:08:49.018.229.000 to 2015-12-03T12:09:19.026.004.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.993326 in 1 packets - #/No: 18/18
      15.013582 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4633028 to 31.4632091
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999532 in 1 packets - #/No: 18/18
      14.9999531 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      07 -> 45.9°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151203_120846_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4815332 to 1.4818222
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9932790  -- OK for SL --
   MIU time from 2015-12-03T12:08:49.018.229.000 to 2015-12-03T12:08:49.018.518.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.993326

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4751012 to 16.4753902
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0135350  -- OK for SL --
   MIU time from 2015-12-03T12:09:04.011.844.000 to 2015-12-03T12:09:04.012.133.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.013582

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4889252 to 31.4892142
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-12-03T12:09:19.025.715.000 to 2015-12-03T12:09:19.026.004.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-12-03T12:08:49.018.088.750
   Max SL corrected packet time: 2015-12-03T12:09:19.025.863.750
   Min SL corrected XO rel time: 0.0182305
   Max SL corrected XO rel time: 30.0260052
   TM values from -17 to 10 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-12-03T12:08:49.018.088.750 (XO: 0.0182305, t=34.0°C)  Rank: 00002
   TM values: Max =     5  Min =   -15  Ave =    -4.66  Std =     2.88
   Min TM value:-15 at an offset of 111.200 usec -> 2015-12-03T12:08:49.018.199.950 (XO: 0.0183417)
   Max TM value:5 at an offset of 5.275 usec -> 2015-12-03T12:08:49.018.094.025 (XO: 0.0182357)
   Max dTM/dt [1/us]::7.90e+01 at an offset of 37.175 usec -> 2015-12-03T12:08:49.018.125.925 (XO: 0.0182676)
   Min dTM/dt [1/us]::-9.10e+01 at an offset of 162.251 usec -> 2015-12-03T12:08:49.018.251.000 (XO: 0.0183927)

 - Block number: 2 starting at 2015-12-03T12:09:04.011.703.750 (XO: 15.0118452, t=34.0°C)  Rank: 00002
   TM values: Max =    10  Min =   -16  Ave =    -4.74  Std =     3.45
   Min TM value:-16 at an offset of 21.725 usec -> 2015-12-03T12:09:04.011.725.475 (XO: 15.0118670)
   Max TM value:10 at an offset of 128.838 usec -> 2015-12-03T12:09:04.011.832.587 (XO: 15.0119741)
   Max dTM/dt [1/us]::9.70e+01 at an offset of 49.375 usec -> 2015-12-03T12:09:04.011.753.125 (XO: 15.0118946)
   Min dTM/dt [1/us]::-8.90e+01 at an offset of 92.800 usec -> 2015-12-03T12:09:04.011.796.550 (XO: 15.0119380)

 - Block number: 3 starting at 2015-12-03T12:09:19.025.574.750 (XO: 30.0257162, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -17  Ave =    -4.58  Std =     3.45
   Min TM value:-17 at an offset of 102.025 usec -> 2015-12-03T12:09:19.025.676.775 (XO: 30.0258182)
   Max TM value:9 at an offset of 84.613 usec -> 2015-12-03T12:09:19.025.659.362 (XO: 30.0258008)
   Max dTM/dt [1/us]::9.40e+01 at an offset of 0.300 usec -> 2015-12-03T12:09:19.025.575.050 (XO: 30.0257165)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 36.288 usec -> 2015-12-03T12:09:19.025.611.037 (XO: 30.0257525)
