-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151130_041922_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.001534 to 30.023803
-each separately from 0s to 30s and from 1534us to 23803us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.009549 in 1 packets - #/No: 18/18
      15.011853 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-30T04:19:25.001.534.000 to 2015-11-30T04:19:55.023.803.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.009549 in 1 packets - #/No: 18/18
      15.011853 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4426766 to 31.4425847
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999540 in 1 packets - #/No: 18/18
      14.9999541 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151130_041922_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4442121 to 1.4445011
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0095030  -- OK for SL --
   MIU time from 2015-11-30T04:19:25.001.534.000 to 2015-11-30T04:19:25.001.823.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.009549

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4540041 to 16.4542931
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0118070  -- OK for SL --
   MIU time from 2015-11-30T04:19:40.011.372.000 to 2015-11-30T04:19:40.011.661.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.011853

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4661001 to 31.4663891
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-30T04:19:55.023.514.000 to 2015-11-30T04:19:55.023.803.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-30T04:19:25.001.393.750
   Max SL corrected packet time: 2015-11-30T04:19:55.023.662.750
   Min SL corrected XO rel time: 0.0015355
   Max SL corrected XO rel time: 30.0238045
   TM values from -12 to 4 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-30T04:19:25.001.393.750 (XO: 0.0015355, t=34.0°C)  Rank: 00002
   TM values: Max =     3  Min =   -12  Ave =    -4.55  Std =     1.96
   Min TM value:-12 at an offset of 9.238 usec -> 2015-11-30T04:19:25.001.402.987 (XO: 0.0015447)
   Max TM value:3 at an offset of 102.650 usec -> 2015-11-30T04:19:25.001.496.400 (XO: 0.0016382)
   Max dTM/dt [1/us]::7.50e+01 at an offset of 169.251 usec -> 2015-11-30T04:19:25.001.563.000 (XO: 0.0017048)
   Min dTM/dt [1/us]::-8.40e+01 at an offset of 190.901 usec -> 2015-11-30T04:19:25.001.584.650 (XO: 0.0017264)

 - Block number: 2 starting at 2015-11-30T04:19:40.011.231.750 (XO: 15.0113735, t=34.0°C)  Rank: 00002
   TM values: Max =     2  Min =   -11  Ave =    -4.64  Std =     1.86
   Min TM value:-11 at an offset of 48.263 usec -> 2015-11-30T04:19:40.011.280.012 (XO: 15.0114218)
   Max TM value:2 at an offset of 4.863 usec -> 2015-11-30T04:19:40.011.236.612 (XO: 15.0113784)
   Max dTM/dt [1/us]::9.20e+01 at an offset of 4.825 usec -> 2015-11-30T04:19:40.011.236.575 (XO: 15.0113784)
   Min dTM/dt [1/us]::-8.20e+01 at an offset of 1.238 usec -> 2015-11-30T04:19:40.011.232.987 (XO: 15.0113748)

 - Block number: 3 starting at 2015-11-30T04:19:55.023.373.750 (XO: 30.0235155, t=34.0°C)  Rank: 00002
   TM values: Max =     4  Min =   -11  Ave =    -4.58  Std =     1.87
   Min TM value:-11 at an offset of 64.713 usec -> 2015-11-30T04:19:55.023.438.462 (XO: 30.0235802)
   Max TM value:4 at an offset of 178.188 usec -> 2015-11-30T04:19:55.023.551.938 (XO: 30.0236937)
   Max dTM/dt [1/us]::8.40e+01 at an offset of 220.413 usec -> 2015-11-30T04:19:55.023.594.163 (XO: 30.0237359)
   Min dTM/dt [1/us]::-8.60e+01 at an offset of 124.575 usec -> 2015-11-30T04:19:55.023.498.325 (XO: 30.0236400)
