-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151126_195852_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.869760 to 41.488348
-each separately from 11s to 41s and from 488059us to 953082us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.082744 in 1 packets - #/No: 18/18
      21.534977 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-26T19:59:06.869.760.000 to 2015-11-26T19:59:36.488.348.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.082744 in 1 packets - #/No: 18/18
      21.534977 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.4740674 to 42.4739750
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999754 in 1 packets - #/No: 18/18
      21.9999322 in 1 packets - #/No: 36/18
Temperature FPGA: 
      09 -> 43.3°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      08 -> 44.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151126_195852_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.3438256 to 13.3441146
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0827190  -- OK for SL --
   MIU time from 2015-11-26T19:59:06.869.760.000 to 2015-11-26T19:59:06.870.049.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.082744

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.4268336 to 21.4271226
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.5349110  -- OK for SL --
   MIU time from 2015-11-26T19:59:14.952.793.000 to 2015-11-26T19:59:14.953.082.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.534977

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.9620336 to 42.9623226
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-26T19:59:36.488.059.000 to 2015-11-26T19:59:36.488.348.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-26T19:59:06.869.619.750
   Max SL corrected packet time: 2015-11-26T19:59:36.488.207.750
   Min SL corrected XO rel time: 11.8697609
   Max SL corrected XO rel time: 41.4883491
   TM values from -28 to 21 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-26T19:59:06.869.619.750 (XO: 11.8697609, t=32.4°C)  Rank: 00009
   TM values: Max =    21  Min =   -28  Ave =    -4.69  Std =     5.19
   Min TM value:-28 at an offset of 256.976 usec -> 2015-11-26T19:59:06.869.876.725 (XO: 11.8700179)
   Max TM value:21 at an offset of 94.663 usec -> 2015-11-26T19:59:06.869.714.412 (XO: 11.8698556)
   Max dTM/dt [1/us]::1.98e+02 at an offset of 191.776 usec -> 2015-11-26T19:59:06.869.811.525 (XO: 11.8699527)
   Min dTM/dt [1/us]::-1.93e+02 at an offset of 139.550 usec -> 2015-11-26T19:59:06.869.759.300 (XO: 11.8699004)

 - Block number: 2 starting at 2015-11-26T19:59:14.952.652.750 (XO: 19.9527938, t=32.4°C)  Rank: 00005
   TM values: Max =    16  Min =   -26  Ave =    -4.73  Std =     4.68
   Min TM value:-26 at an offset of 132.613 usec -> 2015-11-26T19:59:14.952.785.362 (XO: 19.9529264)
   Max TM value:16 at an offset of 120.975 usec -> 2015-11-26T19:59:14.952.773.725 (XO: 19.9529147)
   Max dTM/dt [1/us]::2.16e+02 at an offset of 269.613 usec -> 2015-11-26T19:59:14.952.922.363 (XO: 19.9530634)
   Min dTM/dt [1/us]::-2.27e+02 at an offset of 127.888 usec -> 2015-11-26T19:59:14.952.780.637 (XO: 19.9529216)

 - Block number: 3 starting at 2015-11-26T19:59:36.487.918.750 (XO: 41.4880601, t=32.4°C)  Rank: 00005
   TM values: Max =    15  Min =   -25  Ave =    -4.79  Std =     5.00
   Min TM value:-25 at an offset of 120.175 usec -> 2015-11-26T19:59:36.488.038.925 (XO: 41.4881803)
   Max TM value:15 at an offset of 45.650 usec -> 2015-11-26T19:59:36.487.964.400 (XO: 41.4881058)
   Max dTM/dt [1/us]::2.20e+02 at an offset of 254.676 usec -> 2015-11-26T19:59:36.488.173.425 (XO: 41.4883148)
   Min dTM/dt [1/us]::-2.22e+02 at an offset of 56.088 usec -> 2015-11-26T19:59:36.487.974.837 (XO: 41.4881162)
