-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151126_111010_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.865434 to 32.578558
-each separately from 0s to 32s and from 530416us to 865723us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      27.664693 in 1 packets - #/No: 18/18
      4.047564 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-26T11:10:13.865.434.000 to 2015-11-26T11:10:45.578.558.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      27.664693 in 1 packets - #/No: 18/18
      4.047564 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4007330 to 33.4006338
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      27.9999132 in 1 packets - #/No: 18/18
      3.9999876 in 1 packets - #/No: 36/18
Temperature FPGA: 
      09 -> 43.3°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      08 -> 44.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151126_111010_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2661654 to 2.2664544
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.6646070  -- OK for SL --
   MIU time from 2015-11-26T11:10:13.865.434.000 to 2015-11-26T11:10:13.865.723.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.664693

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.9310614 to 29.9313504
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.0475510  -- OK for SL --
   MIU time from 2015-11-26T11:10:41.530.416.000 to 2015-11-26T11:10:41.530.705.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.047564

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.9789014 to 33.9791904
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-26T11:10:45.578.269.000 to 2015-11-26T11:10:45.578.558.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-26T11:10:13.865.293.750
   Max SL corrected packet time: 2015-11-26T11:10:45.578.417.750
   Min SL corrected XO rel time: 0.8654351
   Max SL corrected XO rel time: 32.5785584
   TM values from -22 to 13 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-26T11:10:13.865.293.750 (XO: 0.8654351, t=32.4°C)  Rank: 00003
   TM values: Max =    13  Min =   -22  Ave =    -4.68  Std =     4.91
   Min TM value:-22 at an offset of 3.450 usec -> 2015-11-26T11:10:13.865.297.200 (XO: 0.8654385)
   Max TM value:13 at an offset of 138.138 usec -> 2015-11-26T11:10:13.865.431.887 (XO: 0.8655732)
   Max dTM/dt [1/us]::1.26e+02 at an offset of 237.013 usec -> 2015-11-26T11:10:13.865.530.763 (XO: 0.8656721)
   Min dTM/dt [1/us]::-1.14e+02 at an offset of 113.825 usec -> 2015-11-26T11:10:13.865.407.575 (XO: 0.8655489)

 - Block number: 2 starting at 2015-11-26T11:10:41.530.275.750 (XO: 28.5304168, t=32.4°C)  Rank: 00004
   TM values: Max =    12  Min =   -21  Ave =    -4.70  Std =     5.12
   Min TM value:-21 at an offset of 0.338 usec -> 2015-11-26T11:10:41.530.276.087 (XO: 28.5304172)
   Max TM value:12 at an offset of 92.575 usec -> 2015-11-26T11:10:41.530.368.325 (XO: 28.5305094)
   Max dTM/dt [1/us]::1.13e+02 at an offset of 197.026 usec -> 2015-11-26T11:10:41.530.472.775 (XO: 28.5306139)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 106.388 usec -> 2015-11-26T11:10:41.530.382.137 (XO: 28.5305232)

 - Block number: 3 starting at 2015-11-26T11:10:45.578.128.750 (XO: 32.5782694, t=32.4°C)  Rank: 00005
   TM values: Max =    11  Min =   -22  Ave =    -4.68  Std =     5.36
   Min TM value:-22 at an offset of 116.400 usec -> 2015-11-26T11:10:45.578.245.150 (XO: 32.5783858)
   Max TM value:11 at an offset of 19.325 usec -> 2015-11-26T11:10:45.578.148.075 (XO: 32.5782887)
   Max dTM/dt [1/us]::9.80e+01 at an offset of 20.513 usec -> 2015-11-26T11:10:45.578.149.262 (XO: 32.5782899)
   Min dTM/dt [1/us]::-1.46e+02 at an offset of 0.000 usec -> 2015-11-26T11:10:45.578.128.750 (XO: 32.5782694)
