-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151125_142123_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 4.281225 to 32.470943
-each separately from 4s to 32s and from 281225us to 758655us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.476852 in 1 packets - #/No: 18/18
      10.711999 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-25T14:21:30.281.225.000 to 2015-11-25T14:21:58.470.943.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.476852 in 1 packets - #/No: 18/18
      10.711999 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.4045751 to 33.4044904
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999486 in 1 packets - #/No: 18/18
      10.9999667 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151125_142123_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.6858008 to 5.6860898
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4767990  -- OK for SL --
   MIU time from 2015-11-25T14:21:30.281.225.000 to 2015-11-25T14:21:30.281.514.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.476852

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.1628888 to 23.1631778
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7119670  -- OK for SL --
   MIU time from 2015-11-25T14:21:47.758.366.000 to 2015-11-25T14:21:47.758.655.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.711999

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.8751448 to 33.8754338
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-25T14:21:58.470.654.000 to 2015-11-25T14:21:58.470.943.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-25T14:21:30.281.084.750
   Max SL corrected packet time: 2015-11-25T14:21:58.470.802.750
   Min SL corrected XO rel time: 4.2812265
   Max SL corrected XO rel time: 32.4709448
   TM values from -23 to 16 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-25T14:21:30.281.084.750 (XO: 4.2812265, t=37.2°C)  Rank: 00003
   TM values: Max =    16  Min =   -20  Ave =    -4.63  Std =     4.42
   Min TM value:-20 at an offset of 7.338 usec -> 2015-11-25T14:21:30.281.092.087 (XO: 4.2812339)
   Max TM value:16 at an offset of 64.863 usec -> 2015-11-25T14:21:30.281.149.612 (XO: 4.2812914)
   Max dTM/dt [1/us]::1.44e+02 at an offset of 305.988 usec -> 2015-11-25T14:21:30.281.390.738 (XO: 4.2815325)
   Min dTM/dt [1/us]::-1.11e+02 at an offset of 64.900 usec -> 2015-11-25T14:21:30.281.149.650 (XO: 4.2812914)

 - Block number: 2 starting at 2015-11-25T14:21:47.758.225.750 (XO: 21.7583675, t=37.2°C)  Rank: 00004
   TM values: Max =    14  Min =   -23  Ave =    -4.52  Std =     5.12
   Min TM value:-23 at an offset of 2.488 usec -> 2015-11-25T14:21:47.758.228.237 (XO: 21.7583699)
   Max TM value:14 at an offset of 130.913 usec -> 2015-11-25T14:21:47.758.356.662 (XO: 21.7584984)
   Max dTM/dt [1/us]::1.07e+02 at an offset of 60.600 usec -> 2015-11-25T14:21:47.758.286.350 (XO: 21.7584281)
   Min dTM/dt [1/us]::-1.03e+02 at an offset of 40.063 usec -> 2015-11-25T14:21:47.758.265.812 (XO: 21.7584075)

 - Block number: 3 starting at 2015-11-25T14:21:58.470.513.750 (XO: 32.4706558, t=37.2°C)  Rank: 00003
   TM values: Max =    13  Min =   -22  Ave =    -4.61  Std =     4.51
   Min TM value:-22 at an offset of 108.000 usec -> 2015-11-25T14:21:58.470.621.750 (XO: 32.4707638)
   Max TM value:13 at an offset of 203.251 usec -> 2015-11-25T14:21:58.470.717.000 (XO: 32.4708591)
   Max dTM/dt [1/us]::1.15e+02 at an offset of 244.001 usec -> 2015-11-25T14:21:58.470.757.750 (XO: 32.4708998)
   Min dTM/dt [1/us]::-1.18e+02 at an offset of 203.288 usec -> 2015-11-25T14:21:58.470.717.038 (XO: 32.4708591)
