-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151124_033340_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.003455 to 44.106082
-each separately from 0s to 44s and from 3455us to 106082us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.009451 in 1 packets - #/No: 18/18
      29.092309 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-24T03:33:43.003.455.000 to 2015-11-24T03:34:27.106.082.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.009451 in 1 packets - #/No: 18/18
      29.092309 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4588279 to 45.4586981
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999557 in 1 packets - #/No: 18/18
      28.9999145 in 1 packets - #/No: 36/18
Temperature FPGA: 
      08 -> 44.6°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      09 -> 43.3°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151124_033340_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4622840 to 1.4625730
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0094070  -- OK for SL --
   MIU time from 2015-11-24T03:33:43.003.455.000 to 2015-11-24T03:33:43.003.744.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.009451

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4719800 to 16.4722690
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.0922230  -- OK for SL --
   MIU time from 2015-11-24T03:33:58.013.195.000 to 2015-11-24T03:33:58.013.484.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.092309

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      41 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.5644920 to 45.5647810
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-24T03:34:27.105.793.000 to 2015-11-24T03:34:27.106.082.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-24T03:33:43.003.314.750
   Max SL corrected packet time: 2015-11-24T03:34:27.105.941.750
   Min SL corrected XO rel time: 0.0034561
   Max SL corrected XO rel time: 44.1060854
   TM values from -95 to 83 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-24T03:33:43.003.314.750 (XO: 0.0034561, t=32.4°C)  Rank: 00002
   TM values: Max =     5  Min =   -16  Ave =    -4.73  Std =     2.82
   Min TM value:-16 at an offset of 123.463 usec -> 2015-11-24T03:33:43.003.438.212 (XO: 0.0035796)
   Max TM value:5 at an offset of 30.000 usec -> 2015-11-24T03:33:43.003.344.750 (XO: 0.0034861)
   Max dTM/dt [1/us]::1.58e+02 at an offset of 123.513 usec -> 2015-11-24T03:33:43.003.438.262 (XO: 0.0035796)
   Min dTM/dt [1/us]::-1.38e+02 at an offset of 230.863 usec -> 2015-11-24T03:33:43.003.545.613 (XO: 0.0036870)

 - Block number: 2 starting at 2015-11-24T03:33:58.013.054.750 (XO: 15.0131964, t=32.4°C)  Rank: 00002
   TM values: Max =     6  Min =   -16  Ave =    -4.76  Std =     3.32
   Min TM value:-16 at an offset of 37.825 usec -> 2015-11-24T03:33:58.013.092.575 (XO: 15.0132343)
   Max TM value:6 at an offset of 40.050 usec -> 2015-11-24T03:33:58.013.094.800 (XO: 15.0132365)
   Max dTM/dt [1/us]::1.75e+02 at an offset of 40.013 usec -> 2015-11-24T03:33:58.013.094.762 (XO: 15.0132365)
   Min dTM/dt [1/us]::-1.59e+02 at an offset of 63.925 usec -> 2015-11-24T03:33:58.013.118.675 (XO: 15.0132604)

 - Block number: 3 starting at 2015-11-24T03:34:27.105.652.750 (XO: 44.1057964, t=32.4°C)  Rank: 00041
   TM values: Max =    83  Min =   -95  Ave =    -4.41  Std =    16.57
   Min TM value:-95 at an offset of 49.600 usec -> 2015-11-24T03:34:27.105.702.350 (XO: 44.1058460)
   Max TM value:83 at an offset of 51.563 usec -> 2015-11-24T03:34:27.105.704.312 (XO: 44.1058480)
   Max dTM/dt [1/us]::4.52e+02 at an offset of 142.938 usec -> 2015-11-24T03:34:27.105.795.687 (XO: 44.1059394)
   Min dTM/dt [1/us]::-6.70e+02 at an offset of 98.175 usec -> 2015-11-24T03:34:27.105.750.925 (XO: 44.1058946)
