-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151123_222742_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.479103 to 30.022354
-each separately from 2s to 30s and from 22065us to 661698us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.182017 in 1 packets - #/No: 18/18
      5.360367 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-23T22:27:47.479.103.000 to 2015-11-23T22:28:15.022.354.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.182017 in 1 packets - #/No: 18/18
      5.360367 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4555938 to 31.4555109
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.9999349 in 1 packets - #/No: 18/18
      5.9999822 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      08 -> 44.6°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151123_222742_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.9346969 to 3.9349859
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1819510  -- OK for SL --
   MIU time from 2015-11-23T22:27:47.479.103.000 to 2015-11-23T22:27:47.479.392.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.182017

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      51 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.1169369 to 26.1172259
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.3603510  -- OK for SL --
   MIU time from 2015-11-23T22:28:09.661.409.000 to 2015-11-23T22:28:09.661.698.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.360367

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4775769 to 31.4778659
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-23T22:28:15.022.065.000 to 2015-11-23T22:28:15.022.354.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-23T22:27:47.478.962.750
   Max SL corrected packet time: 2015-11-23T22:28:15.022.213.750
   Min SL corrected XO rel time: 2.4791045
   Max SL corrected XO rel time: 30.0223551
   TM values from -312 to 185 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-23T22:27:47.478.962.750 (XO: 2.4791045, t=32.4°C)  Rank: 00003
   TM values: Max =    11  Min =   -20  Ave =    -4.65  Std =     3.99
   Min TM value:-20 at an offset of 122.888 usec -> 2015-11-23T22:27:47.479.085.637 (XO: 2.4792274)
   Max TM value:11 at an offset of 134.213 usec -> 2015-11-23T22:27:47.479.096.962 (XO: 2.4792387)
   Max dTM/dt [1/us]::1.42e+02 at an offset of 247.801 usec -> 2015-11-23T22:27:47.479.210.550 (XO: 2.4793523)
   Min dTM/dt [1/us]::-1.43e+02 at an offset of 28.588 usec -> 2015-11-23T22:27:47.478.991.337 (XO: 2.4791331)

 - Block number: 2 starting at 2015-11-23T22:28:09.661.268.750 (XO: 24.6614102, t=32.4°C)  Rank: 00051
   TM values: Max =   185  Min =  -312  Ave =    -4.64  Std =    20.76
   Min TM value:-312 at an offset of 112.588 usec -> 2015-11-23T22:28:09.661.381.337 (XO: 24.6615228)
   Max TM value:185 at an offset of 112.325 usec -> 2015-11-23T22:28:09.661.381.075 (XO: 24.6615225)
   Max dTM/dt [1/us]::2.14e+03 at an offset of 112.800 usec -> 2015-11-23T22:28:09.661.381.550 (XO: 24.6615230)
   Min dTM/dt [1/us]::-3.75e+03 at an offset of 112.525 usec -> 2015-11-23T22:28:09.661.381.275 (XO: 24.6615227)

 - Block number: 3 starting at 2015-11-23T22:28:15.021.924.750 (XO: 30.0220661, t=32.4°C)  Rank: 00002
   TM values: Max =     7  Min =   -18  Ave =    -4.76  Std =     3.67
   Min TM value:-18 at an offset of 135.413 usec -> 2015-11-23T22:28:15.022.060.162 (XO: 30.0222015)
   Max TM value:7 at an offset of 6.575 usec -> 2015-11-23T22:28:15.021.931.325 (XO: 30.0220726)
   Max dTM/dt [1/us]::1.53e+02 at an offset of 136.263 usec -> 2015-11-23T22:28:15.022.061.012 (XO: 30.0222023)
   Min dTM/dt [1/us]::-1.52e+02 at an offset of 221.213 usec -> 2015-11-23T22:28:15.022.145.963 (XO: 30.0222873)
