-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151121_222214_1229.dat, 36720  packets found.
APID:
      1229 (EW) in 36720 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36720 packets (all packets)
Packet no from 1 to 36720
   Differences of packet numbers in neigboring packets: 
      1 in 36719 packets (all packets)
Total number of packets: 
      36720 in 36720 packets (all packets)
Hz and MHz counters from 336.014726 to 1377.995707
-each separately from 336s to 1377s and from 14726us to 995707us.
   Differences in neigboring packets: 
      0.000000 in 36710 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      0.946767 in 2 packets - #/No: 3468/3468, 18768/18768
      1041.034214 in 2 packets - #/No: 5916/5916, 22440/22440
      -1041.980981 in 2 packets - #/No: 7752/7752, 28968/28968
      1041.980981 in 1 packets - #/No: 8160/8160
      -1041.034214 in 1 packets - #/No: 12036/12036
      -0.946767 in 1 packets - #/No: 18156/18156
MIU packet time from 2015-11-21T22:27:53.014.726.000 to 2015-11-21T22:45:14.995.707.000
   Differences in neigboring packets: 
      0.000000 in 36710 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      0.946767 in 2 packets - #/No: 3468/3468, 18768/18768
      1041.034214 in 2 packets - #/No: 5916/5916, 22440/22440
      -1041.980981 in 2 packets - #/No: 7752/7752, 28968/28968
      1041.980981 in 1 packets - #/No: 8160/8160
      -1041.034214 in 1 packets - #/No: 12036/12036
      -0.946767 in 1 packets - #/No: 18156/18156
MIU orbit number: 
      60000 in 36720 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36720 packets (all packets)
MIU orbit information / number of events: 
      0 in 36720 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36720 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36720 packets (all packets)
FPGA configuration version: 
      211 in 36720 packets (all packets)
10 MHz counter at the last 1Hz pulse from 337.3954125 to 1378.3923741
   Differences in neigboring packets: 
      0.0000000 in 36713 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1040.9969616 in 3 packets - #/No: 5916/5916, 8160/8160, 22440/22440
      -1040.9969616 in 3 packets - #/No: 7752/7752, 12036/12036, 28968/28968
Temperature FPGA: 
      08 -> 44.6°C in 36720 packets (all packets)
Temperature XO: 
      BE -> 32.4°C in 36720 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 36720 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 36720 packets (all packets)
Number of external events: 
      2 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      3 in 12240 packets - #/No: 3469/3469, 3470/3470, 3471/3471, 3472/3472, 3473/3473, 3474/3474, 3475/3475, 3476/3476, 3477/3477, 3478/3478, 3479/3479, 3480/3480, 3481/3481, 3482/3482, 3483/3483, 3484/3484, 3485/3485, 3486/3486, ...
      1 in 12240 packets - #/No: 5917/5917, 5918/5918, 5919/5919, 5920/5920, 5921/5921, 5922/5922, 5923/5923, 5924/5924, 5925/5925, 5926/5926, 5927/5927, 5928/5928, 5929/5929, 5930/5930, 5931/5931, 5932/5932, 5933/5933, 5934/5934, ...
Number of generated alerts: 
      2 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      3 in 12240 packets - #/No: 3469/3469, 3470/3470, 3471/3471, 3472/3472, 3473/3473, 3474/3474, 3475/3475, 3476/3476, 3477/3477, 3478/3478, 3479/3479, 3480/3480, 3481/3481, 3482/3482, 3483/3483, 3484/3484, 3485/3485, 3486/3486, ...
      1 in 12240 packets - #/No: 5917/5917, 5918/5918, 5919/5919, 5920/5920, 5921/5921, 5922/5922, 5923/5923, 5924/5924, 5925/5925, 5926/5926, 5927/5927, 5928/5928, 5929/5929, 5930/5930, 5931/5931, 5932/5932, 5933/5933, 5934/5934, ...
Survey memory error: 
      0 in 36720 packets (all packets)
Event memory error: 
      0 in 36720 packets (all packets)
Configuration error: 
      0 in 36720 packets (all packets)
Zero: 
      0 in 36720 packets (all packets)
Radio gain: 
      0 in 36720 packets (all packets)
Radio bandwidth: 
      0 in 36720 packets (all packets)
Digital gain: 
      0 in 36720 packets (all packets)
Analog gain: 
      0 in 36720 packets (all packets)

EVENT DATA READER
-- File 20151121_222214_1229.dat, 180 EW pages found. Extra packets: 0.
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      2 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      3 in 12240 packets - #/No: 3469/3469, 3470/3470, 3471/3471, 3472/3472, 3473/3473, 3474/3474, 3475/3475, 3476/3476, 3477/3477, 3478/3478, 3479/3479, 3480/3480, 3481/3481, 3482/3482, 3483/3483, 3484/3484, 3485/3485, 3486/3486, ...
      1 in 12240 packets - #/No: 5917/5917, 5918/5918, 5919/5919, 5920/5920, 5921/5921, 5922/5922, 5923/5923, 5924/5924, 5925/5925, 5926/5926, 5927/5927, 5928/5928, 5929/5929, 5930/5930, 5931/5931, 5932/5932, 5933/5933, 5934/5934, ...
   10 MHz counter from 90.8424024 to 338.5094782
      Difference between the neighboring packets: 
      0.0000170 in 36710 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      0.9225050 in 1 packets - #/No: 3468/3468
      181.8296538 in 1 packets - #/No: 5916/5916
      -182.8249358 in 1 packets - #/No: 7752/7752
      182.8180338 in 1 packets - #/No: 8160/8160
      -182.1348038 in 1 packets - #/No: 12036/12036
      -0.8531110 in 1 packets - #/No: 18156/18156
      0.8427410 in 1 packets - #/No: 18768/18768
      181.9683738 in 1 packets - #/No: 22440/22440
      -183.1925438 in 1 packets - #/No: 28968/28968
   Memory page number from 0 to 237
      Difference between the neighboring packets: 
      0 in 36540 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 175 packets - #/No: 204/204, 408/408, 612/612, 816/816, 1020/1020, 1224/1224, 1428/1428, 1632/1632, 1836/1836, 2040/2040, 2244/2244, 2448/2448, 2652/2652, 2856/2856, 3060/3060, 3264/3264, 3672/3672, 3876/3876, ...
      3 in 1 packets - #/No: 3468/3468
      31 in 1 packets - #/No: 5916/5916
      19 in 1 packets - #/No: 8160/8160
      9 in 1 packets - #/No: 28968/28968
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 36540 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 170 packets - #/No: 204/204, 408/408, 612/612, 816/816, 1020/1020, 1224/1224, 1428/1428, 1632/1632, 1836/1836, 2040/2040, 2244/2244, 2448/2448, 2652/2652, 2856/2856, 3060/3060, 3264/3264, 3672/3672, 3876/3876, ...
      -6 in 1 packets - #/No: 3468/3468
      -59 in 2 packets - #/No: 5916/5916, 28968/28968
      47 in 1 packets - #/No: 7752/7752
      -47 in 1 packets - #/No: 8160/8160
      -27 in 1 packets - #/No: 12036/12036
      28 in 1 packets - #/No: 18156/18156
      -29 in 1 packets - #/No: 18768/18768
      -19 in 1 packets - #/No: 22440/22440
   MIU event flag 
      1 in 36720 packets (all packets)
   Time triggered event flag 
      0 in 36720 packets (all packets)
   Min EW event (1st sample) abs. time: 2015-11-21T22:27:52.959.237.590
   Min EW header abs. time            : 2015-11-21T22:27:53.014.726.000
   Max EW event (1st sample) abs. time: 2015-11-21T22:45:14.940.218.590
   Max EW header abs. time            : 2015-11-21T22:45:14.995.707.000
   TM values from -2048 to 1784 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      69 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      103 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      104 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      105 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      106 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      107 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      108 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      160 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      161 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      162 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      163 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      164 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      165 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      166 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      167 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      168 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      169 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      170 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      171 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      172 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      173 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      174 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      175 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      176 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      177 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      178 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      179 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      180 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      181 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      182 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      183 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      184 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      185 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      186 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      187 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      188 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      189 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      190 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      191 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 60  ordered packets: 12240
   TM values from -2048 to 1784 median:-5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=1377 C1MHz=0.995707
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=90.8424024
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2015-11-21T22:45:14.940.218.590
      (header abs. time: 2015-11-21T22:45:14.995.707.000)
      Event 1st sample XO time (corr. sec. from XP start):              89.446274344
      Event header XO time (corr. sec. from XP start)    :              89.446274594
      Sampling frequency correction factor for this event:     0.999997100

 Event no. 2 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      200 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      201 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      202 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      203 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      204 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      205 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      206 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      207 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      208 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      209 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      210 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      211 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      212 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      213 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      214 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      215 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      216 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      217 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      218 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      219 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      220 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      221 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      222 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      223 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      224 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      225 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      226 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      227 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      228 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      229 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      230 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      231 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      232 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      233 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      234 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      235 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      236 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      237 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      139 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      140 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      141 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 60  ordered packets: 12240
   TM values from -872 to 603 median:-5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=336 C1MHz=0.014726
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=337.3546512
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2015-11-21T22:27:52.959.237.590
      (header abs. time: 2015-11-21T22:27:53.014.726.000)
      Event 1st sample XO time (corr. sec. from XP start):             335.959238032
      Event header XO time (corr. sec. from XP start)    :             335.959238282
      Sampling frequency correction factor for this event:     0.999997100

 Event no. 3 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      109 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      110 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      111 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      112 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      113 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      114 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      115 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      116 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      117 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      118 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      119 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      120 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      121 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      122 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      123 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      124 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      125 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      126 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      127 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      128 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      129 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      130 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      131 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      132 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      133 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      134 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      135 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      136 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      137 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      138 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      142 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      143 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      144 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      145 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      146 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      147 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      148 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      149 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      150 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      151 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      152 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      153 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      154 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      155 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      156 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      157 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      158 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      159 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 60  ordered packets: 12240
   TM values from -272 to 744 median:-5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=336 C1MHz=0.961493
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=338.3014152
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2015-11-21T22:27:53.906.004.590
      (header abs. time: 2015-11-21T22:27:53.961.493.000)
      Event 1st sample XO time (corr. sec. from XP start):             336.906004777
      Event header XO time (corr. sec. from XP start)    :             336.906005027
      Sampling frequency correction factor for this event:     0.999997100
--
Extremes in separate events:

 - Event number: 1 starting at 2015-11-21T22:45:14.940.218.590 (XO: 89.4462743)
   Min TM value:-2048 at an offset of 14402.992 usec -> 2015-11-21T22:45:14.954.621.581 (XO: 89.4606773)
   Max TM value:1784 at an offset of 93461.271 usec -> 2015-11-21T22:45:15.033.679.861 (XO: 89.5397356)
   Max dTM/dt [1/us]:2.18e+04 at an offset of 93461.221 usec -> 2015-11-21T22:45:15.033.679.811 (XO: 89.5397356)
   Min dTM/dt [1/us]:-2.56e+04 at an offset of 59878.036 usec -> 2015-11-21T22:45:15.000.096.626 (XO: 89.5061524)

 - Event number: 2 starting at 2015-11-21T22:27:52.959.237.590 (XO: 335.9592380)
   Min TM value:-872 at an offset of 55292.810 usec -> 2015-11-21T22:27:53.014.530.400 (XO: 336.0145308)
   Max TM value:603 at an offset of 55300.098 usec -> 2015-11-21T22:27:53.014.537.687 (XO: 336.0145381)
   Max dTM/dt [1/us]:4.40e+03 at an offset of 55293.223 usec -> 2015-11-21T22:27:53.014.530.812 (XO: 336.0145313)
   Min dTM/dt [1/us]:-3.47e+03 at an offset of 55292.760 usec -> 2015-11-21T22:27:53.014.530.350 (XO: 336.0145308)

 - Event number: 3 starting at 2015-11-21T22:27:53.906.004.590 (XO: 336.9060048)
   Min TM value:-272 at an offset of 53840.281 usec -> 2015-11-21T22:27:53.959.844.871 (XO: 336.9598451)
   Max TM value:744 at an offset of 53825.456 usec -> 2015-11-21T22:27:53.959.830.046 (XO: 336.9598302)
   Max dTM/dt [1/us]:3.52e+03 at an offset of 53825.419 usec -> 2015-11-21T22:27:53.959.830.008 (XO: 336.9598302)
   Min dTM/dt [1/us]:-3.28e+03 at an offset of 53825.544 usec -> 2015-11-21T22:27:53.959.830.133 (XO: 336.9598303)
