-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151121_011438_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.204720 to 31.897624
-each separately from 7s to 31s and from 204720us to 914058us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.708760 in 1 packets - #/No: 18/18
      15.983277 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-21T01:14:48.204.720.000 to 2015-11-21T01:15:12.897.624.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.708760 in 1 packets - #/No: 18/18
      15.983277 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4213200 to 32.4212504
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999768 in 1 packets - #/No: 18/18
      15.9999536 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151121_011438_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6260403 to 8.6263293
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7087350  -- OK for SL --
   MIU time from 2015-11-21T01:14:48.204.720.000 to 2015-11-21T01:14:48.205.009.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.708760

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3350643 to 17.3353533
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9832310  -- OK for SL --
   MIU time from 2015-11-21T01:14:56.913.769.000 to 2015-11-21T01:14:56.914.058.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.983277

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.3185843 to 33.3188733
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-21T01:15:12.897.335.000 to 2015-11-21T01:15:12.897.624.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-21T01:14:48.204.579.750
   Max SL corrected packet time: 2015-11-21T01:15:12.897.483.750
   Min SL corrected XO rel time: 7.2047209
   Max SL corrected XO rel time: 31.8976255
   TM values from -21 to 11 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-21T01:14:48.204.579.750 (XO: 7.2047209, t=34.0°C)  Rank: 00004
   TM values: Max =    11  Min =   -21  Ave =    -4.72  Std =     4.46
   Min TM value:-21 at an offset of 196.913 usec -> 2015-11-21T01:14:48.204.776.663 (XO: 7.2049178)
   Max TM value:11 at an offset of 34.963 usec -> 2015-11-21T01:14:48.204.614.712 (XO: 7.2047559)
   Max dTM/dt [1/us]::2.39e+02 at an offset of 114.638 usec -> 2015-11-21T01:14:48.204.694.387 (XO: 7.2048355)
   Min dTM/dt [1/us]::-2.50e+02 at an offset of 198.363 usec -> 2015-11-21T01:14:48.204.778.113 (XO: 7.2049193)

 - Block number: 2 starting at 2015-11-21T01:14:56.913.628.750 (XO: 15.9137701, t=34.0°C)  Rank: 00005
   TM values: Max =    11  Min =   -21  Ave =    -4.59  Std =     4.85
   Min TM value:-21 at an offset of 111.138 usec -> 2015-11-21T01:14:56.913.739.887 (XO: 15.9138813)
   Max TM value:11 at an offset of 110.388 usec -> 2015-11-21T01:14:56.913.739.137 (XO: 15.9138805)
   Max dTM/dt [1/us]::2.81e+02 at an offset of 119.763 usec -> 2015-11-21T01:14:56.913.748.512 (XO: 15.9138899)
   Min dTM/dt [1/us]::-2.95e+02 at an offset of 111.100 usec -> 2015-11-21T01:14:56.913.739.850 (XO: 15.9138813)

 - Block number: 3 starting at 2015-11-21T01:15:12.897.194.750 (XO: 31.8973365, t=34.0°C)  Rank: 00003
   TM values: Max =     9  Min =   -18  Ave =    -4.72  Std =     4.02
   Min TM value:-18 at an offset of 88.238 usec -> 2015-11-21T01:15:12.897.282.987 (XO: 31.8974247)
   Max TM value:9 at an offset of 78.088 usec -> 2015-11-21T01:15:12.897.272.837 (XO: 31.8974146)
   Max dTM/dt [1/us]::2.12e+02 at an offset of 86.500 usec -> 2015-11-21T01:15:12.897.281.250 (XO: 31.8974230)
   Min dTM/dt [1/us]::-2.05e+02 at an offset of 78.125 usec -> 2015-11-21T01:15:12.897.272.875 (XO: 31.8974146)
