-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151120_193232_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.675063 to 41.549967
-each separately from 0s to 41s and from 549678us to 675352us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.897017 in 1 packets - #/No: 18/18
      20.977020 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-20T19:32:35.675.063.000 to 2015-11-20T19:33:16.549.967.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.897017 in 1 packets - #/No: 18/18
      20.977020 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4100710 to 42.4099520
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.9999419 in 1 packets - #/No: 18/18
      20.9999391 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151120_193232_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0851335 to 2.0854225
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.8969590  -- OK for SL --
   MIU time from 2015-11-20T19:32:35.675.063.000 to 2015-11-20T19:32:35.675.352.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.897017

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.9823815 to 21.9826705
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9769590  -- OK for SL --
   MIU time from 2015-11-20T19:32:55.572.369.000 to 2015-11-20T19:32:55.572.658.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.977020

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.9596295 to 42.9599185
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-20T19:33:16.549.678.000 to 2015-11-20T19:33:16.549.967.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-20T19:32:35.674.922.750
   Max SL corrected packet time: 2015-11-20T19:33:16.549.826.750
   Min SL corrected XO rel time: 0.6750645
   Max SL corrected XO rel time: 41.5499681
   TM values from -40 to 34 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-20T19:32:35.674.922.750 (XO: 0.6750645, t=35.6°C)  Rank: 00010
   TM values: Max =    34  Min =   -40  Ave =    -4.65  Std =    13.31
   Min TM value:-40 at an offset of 116.963 usec -> 2015-11-20T19:32:35.675.039.712 (XO: 0.6751815)
   Max TM value:34 at an offset of 215.201 usec -> 2015-11-20T19:32:35.675.137.950 (XO: 0.6752797)
   Max dTM/dt [1/us]::6.31e+02 at an offset of 124.863 usec -> 2015-11-20T19:32:35.675.047.612 (XO: 0.6751894)
   Min dTM/dt [1/us]::-6.28e+02 at an offset of 224.451 usec -> 2015-11-20T19:32:35.675.147.200 (XO: 0.6752890)

 - Block number: 2 starting at 2015-11-20T19:32:55.572.228.750 (XO: 20.5723703, t=35.6°C)  Rank: 00009
   TM values: Max =    27  Min =   -36  Ave =    -4.69  Std =     8.59
   Min TM value:-36 at an offset of 13.600 usec -> 2015-11-20T19:32:55.572.242.350 (XO: 20.5723839)
   Max TM value:27 at an offset of 14.363 usec -> 2015-11-20T19:32:55.572.243.112 (XO: 20.5723846)
   Max dTM/dt [1/us]::5.17e+02 at an offset of 12.150 usec -> 2015-11-20T19:32:55.572.240.900 (XO: 20.5723824)
   Min dTM/dt [1/us]::-5.28e+02 at an offset of 13.563 usec -> 2015-11-20T19:32:55.572.242.312 (XO: 20.5723838)

 - Block number: 3 starting at 2015-11-20T19:33:16.549.537.750 (XO: 41.5496791, t=35.6°C)  Rank: 00008
   TM values: Max =    26  Min =   -38  Ave =    -4.77  Std =     8.58
   Min TM value:-38 at an offset of 74.613 usec -> 2015-11-20T19:33:16.549.612.362 (XO: 41.5497537)
   Max TM value:26 at an offset of 75.363 usec -> 2015-11-20T19:33:16.549.613.112 (XO: 41.5497545)
   Max dTM/dt [1/us]::5.72e+02 at an offset of 75.475 usec -> 2015-11-20T19:33:16.549.613.225 (XO: 41.5497546)
   Min dTM/dt [1/us]::-5.80e+02 at an offset of 75.400 usec -> 2015-11-20T19:33:16.549.613.150 (XO: 41.5497545)
