-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151120_073714_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 14.885226 to 30.417496
-each separately from 14s to 30s and from 417207us to 885515us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.542505 in 1 packets - #/No: 18/18
      11.988898 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-20T07:37:31.885.226.000 to 2015-11-20T07:37:47.417.496.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.542505 in 1 packets - #/No: 18/18
      11.988898 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.3934388 to 31.3933918
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.9999882 in 1 packets - #/No: 18/18
      11.9999648 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      08 -> 44.6°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BE -> 32.4°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      BF -> 34.0°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151120_073714_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.2786630 to 16.2789520
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.5424950  -- OK for SL --
   MIU time from 2015-11-20T07:37:31.885.226.000 to 2015-11-20T07:37:31.885.515.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.542505

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.8214470 to 19.8217360
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9888630  -- OK for SL --
   MIU time from 2015-11-20T07:37:35.428.020.000 to 2015-11-20T07:37:35.428.309.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.988898

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.8105990 to 31.8108880
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-20T07:37:47.417.207.000 to 2015-11-20T07:37:47.417.496.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-20T07:37:31.885.085.750
   Max SL corrected packet time: 2015-11-20T07:37:47.417.355.750
   Min SL corrected XO rel time: 14.8852269
   Max SL corrected XO rel time: 30.4174975
   TM values from -30 to 21 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-20T07:37:31.885.085.750 (XO: 14.8852269, t=32.4°C)  Rank: 00007
   TM values: Max =    16  Min =   -26  Ave =    -4.76  Std =     6.16
   Min TM value:-26 at an offset of 0.775 usec -> 2015-11-20T07:37:31.885.086.525 (XO: 14.8852276)
   Max TM value:16 at an offset of 1.238 usec -> 2015-11-20T07:37:31.885.086.987 (XO: 14.8852281)
   Max dTM/dt [1/us]::2.47e+02 at an offset of 0.013 usec -> 2015-11-20T07:37:31.885.085.762 (XO: 14.8852269)
   Min dTM/dt [1/us]::-9.60e+01 at an offset of 10.763 usec -> 2015-11-20T07:37:31.885.096.512 (XO: 14.8852376)

 - Block number: 2 starting at 2015-11-20T07:37:35.427.879.750 (XO: 18.4280212, t=32.4°C)  Rank: 00009
   TM values: Max =    21  Min =   -29  Ave =    -4.71  Std =     7.09
   Min TM value:-29 at an offset of 134.825 usec -> 2015-11-20T07:37:35.428.014.575 (XO: 18.4281561)
   Max TM value:21 at an offset of 133.938 usec -> 2015-11-20T07:37:35.428.013.687 (XO: 18.4281552)
   Max dTM/dt [1/us]::1.04e+02 at an offset of 36.150 usec -> 2015-11-20T07:37:35.427.915.900 (XO: 18.4280574)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 133.025 usec -> 2015-11-20T07:37:35.428.012.775 (XO: 18.4281543)

 - Block number: 3 starting at 2015-11-20T07:37:47.417.066.750 (XO: 30.4172085, t=32.4°C)  Rank: 00006
   TM values: Max =    21  Min =   -30  Ave =    -4.68  Std =     9.30
   Min TM value:-30 at an offset of 286.726 usec -> 2015-11-20T07:37:47.417.353.475 (XO: 30.4174952)
   Max TM value:21 at an offset of 243.326 usec -> 2015-11-20T07:37:47.417.310.075 (XO: 30.4174518)
   Max dTM/dt [1/us]::1.52e+02 at an offset of 0.013 usec -> 2015-11-20T07:37:47.417.066.762 (XO: 30.4172085)
   Min dTM/dt [1/us]::-1.80e+02 at an offset of 305.988 usec -> 2015-11-20T07:37:47.417.372.738 (XO: 30.4175144)
