-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151115_130408_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.707651 to 30.523480
-each separately from 1s to 30s and from 523191us to 744237us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.036008 in 1 packets - #/No: 18/18
      14.778954 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-15T13:04:12.707.651.000 to 2015-11-15T13:04:41.523.480.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.036008 in 1 packets - #/No: 18/18
      14.778954 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4518635 to 31.4517792
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999593 in 1 packets - #/No: 18/18
      14.9999564 in 1 packets - #/No: 36/18
Temperature FPGA: 
      08 -> 44.6°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151115_130408_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.1595137 to 3.1598027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0359670  -- OK for SL --
   MIU time from 2015-11-15T13:04:12.707.651.000 to 2015-11-15T13:04:12.707.940.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.036008

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1957697 to 17.1960587
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7789110  -- OK for SL --
   MIU time from 2015-11-15T13:04:26.743.948.000 to 2015-11-15T13:04:26.744.237.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.778954

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9749697 to 31.9752587
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-15T13:04:41.523.191.000 to 2015-11-15T13:04:41.523.480.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-15T13:04:12.707.510.750
   Max SL corrected packet time: 2015-11-15T13:04:41.523.339.750
   Min SL corrected XO rel time: 1.7076523
   Max SL corrected XO rel time: 30.5234810
   TM values from -23 to 13 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-15T13:04:12.707.510.750 (XO: 1.7076523, t=34.0°C)  Rank: 00003
   TM values: Max =    13  Min =   -23  Ave =    -4.59  Std =     4.89
   Min TM value:-23 at an offset of 231.926 usec -> 2015-11-15T13:04:12.707.742.675 (XO: 1.7078842)
   Max TM value:13 at an offset of 231.038 usec -> 2015-11-15T13:04:12.707.741.788 (XO: 1.7078833)
   Max dTM/dt [1/us]::1.11e+02 at an offset of 128.538 usec -> 2015-11-15T13:04:12.707.639.287 (XO: 1.7077808)
   Min dTM/dt [1/us]::-1.37e+02 at an offset of 305.988 usec -> 2015-11-15T13:04:12.707.816.738 (XO: 1.7079582)

 - Block number: 2 starting at 2015-11-15T13:04:26.743.807.750 (XO: 15.7439491, t=34.0°C)  Rank: 00003
   TM values: Max =    10  Min =   -22  Ave =    -4.62  Std =     4.11
   Min TM value:-22 at an offset of 88.900 usec -> 2015-11-15T13:04:26.743.896.650 (XO: 15.7440380)
   Max TM value:10 at an offset of 111.225 usec -> 2015-11-15T13:04:26.743.918.975 (XO: 15.7440604)
   Max dTM/dt [1/us]::9.30e+01 at an offset of 88.813 usec -> 2015-11-15T13:04:26.743.896.562 (XO: 15.7440379)
   Min dTM/dt [1/us]::-9.90e+01 at an offset of 53.250 usec -> 2015-11-15T13:04:26.743.861.000 (XO: 15.7440024)

 - Block number: 3 starting at 2015-11-15T13:04:41.523.050.750 (XO: 30.5231920, t=34.0°C)  Rank: 00004
   TM values: Max =     8  Min =   -18  Ave =    -4.65  Std =     3.41
   Min TM value:-18 at an offset of 45.225 usec -> 2015-11-15T13:04:41.523.095.975 (XO: 30.5232372)
   Max TM value:8 at an offset of 9.550 usec -> 2015-11-15T13:04:41.523.060.300 (XO: 30.5232016)
   Max dTM/dt [1/us]::1.00e+02 at an offset of 262.076 usec -> 2015-11-15T13:04:41.523.312.825 (XO: 30.5234541)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 282.263 usec -> 2015-11-15T13:04:41.523.333.013 (XO: 30.5234743)
