-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151110_193314_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 9.146900 to 34.556639
-each separately from 9s to 34s and from 146900us to 949961us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.802483 in 1 packets - #/No: 18/18
      18.606389 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-10T19:33:26.146.900.000 to 2015-11-10T19:33:51.556.639.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.802483 in 1 packets - #/No: 18/18
      18.606389 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4133471 to 35.4132747
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.9999826 in 1 packets - #/No: 18/18
      18.9999450 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 54 packets (all packets)
Temperature XO: 
      BE -> 32.4°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      BF -> 34.0°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151110_193314_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.5602483 to 10.5605373
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.8024630  -- OK for SL --
   MIU time from 2015-11-10T19:33:26.146.900.000 to 2015-11-10T19:33:26.147.189.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.802483

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3630003 to 17.3632893
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.6063350  -- OK for SL --
   MIU time from 2015-11-10T19:33:32.949.672.000 to 2015-11-10T19:33:32.949.961.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.606389

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.9696243 to 35.9699133
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-10T19:33:51.556.350.000 to 2015-11-10T19:33:51.556.639.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-10T19:33:26.146.759.750
   Max SL corrected packet time: 2015-11-10T19:33:51.556.498.750
   Min SL corrected XO rel time: 9.1469016
   Max SL corrected XO rel time: 34.5566402
   TM values from -28 to 19 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-10T19:33:26.146.759.750 (XO: 9.1469016, t=32.4°C)  Rank: 00004
   TM values: Max =    17  Min =   -21  Ave =    -4.59  Std =     5.00
   Min TM value:-21 at an offset of 118.438 usec -> 2015-11-10T19:33:26.146.878.187 (XO: 9.1470201)
   Max TM value:17 at an offset of 116.338 usec -> 2015-11-10T19:33:26.146.876.087 (XO: 9.1470180)
   Max dTM/dt [1/us]::2.96e+02 at an offset of 116.300 usec -> 2015-11-10T19:33:26.146.876.050 (XO: 9.1470179)
   Min dTM/dt [1/us]::-3.00e+02 at an offset of 118.400 usec -> 2015-11-10T19:33:26.146.878.150 (XO: 9.1470200)

 - Block number: 2 starting at 2015-11-10T19:33:32.949.531.750 (XO: 15.9496733, t=32.4°C)  Rank: 00004
   TM values: Max =    19  Min =   -28  Ave =    -4.67  Std =     5.61
   Min TM value:-28 at an offset of 98.888 usec -> 2015-11-10T19:33:32.949.630.637 (XO: 15.9497721)
   Max TM value:19 at an offset of 305.313 usec -> 2015-11-10T19:33:32.949.837.063 (XO: 15.9499786)
   Max dTM/dt [1/us]::3.56e+02 at an offset of 97.438 usec -> 2015-11-10T19:33:32.949.629.187 (XO: 15.9497707)
   Min dTM/dt [1/us]::-3.20e+02 at an offset of 98.025 usec -> 2015-11-10T19:33:32.949.629.775 (XO: 15.9497713)

 - Block number: 3 starting at 2015-11-10T19:33:51.556.209.750 (XO: 34.5563512, t=32.4°C)  Rank: 00005
   TM values: Max =    18  Min =   -27  Ave =    -4.66  Std =     5.56
   Min TM value:-27 at an offset of 112.125 usec -> 2015-11-10T19:33:51.556.321.875 (XO: 34.5564633)
   Max TM value:18 at an offset of 111.225 usec -> 2015-11-10T19:33:51.556.320.975 (XO: 34.5564624)
   Max dTM/dt [1/us]::3.50e+02 at an offset of 111.850 usec -> 2015-11-10T19:33:51.556.321.600 (XO: 34.5564631)
   Min dTM/dt [1/us]::-3.83e+02 at an offset of 111.263 usec -> 2015-11-10T19:33:51.556.321.012 (XO: 34.5564625)
