-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151109_044044_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.563114 to 30.023312
-each separately from 5s to 30s and from 23023us to 563403us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.784834 in 1 packets - #/No: 18/18
      0.674497 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-09T04:40:52.563.114.000 to 2015-11-09T04:41:17.023.312.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.784834 in 1 packets - #/No: 18/18
      0.674497 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4372088 to 31.4371380
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999321 in 1 packets - #/No: 18/18
      0.9999971 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151109_044044_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.0003226 to 7.0006116
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.7847670  -- OK for SL --
   MIU time from 2015-11-09T04:40:52.563.114.000 to 2015-11-09T04:40:52.563.403.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.784834

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.7853786 to 30.7856676
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.6744950  -- OK for SL --
   MIU time from 2015-11-09T04:41:16.348.237.000 to 2015-11-09T04:41:16.348.526.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.674497

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4601626 to 31.4604516
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-09T04:41:17.023.023.000 to 2015-11-09T04:41:17.023.312.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-09T04:40:52.562.973.750
   Max SL corrected packet time: 2015-11-09T04:41:17.023.171.750
   Min SL corrected XO rel time: 5.5631154
   Max SL corrected XO rel time: 30.0233137
   TM values from -12 to 4 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-09T04:40:52.562.973.750 (XO: 5.5631154, t=35.6°C)  Rank: 00003
   TM values: Max =     4  Min =   -11  Ave =    -4.58  Std =     1.77
   Min TM value:-11 at an offset of 260.326 usec -> 2015-11-09T04:40:52.563.234.075 (XO: 5.5633758)
   Max TM value:4 at an offset of 198.901 usec -> 2015-11-09T04:40:52.563.172.650 (XO: 5.5633143)
   Max dTM/dt [1/us]::6.30e+01 at an offset of 42.538 usec -> 2015-11-09T04:40:52.563.016.287 (XO: 5.5631580)
   Min dTM/dt [1/us]::-6.20e+01 at an offset of 42.625 usec -> 2015-11-09T04:40:52.563.016.375 (XO: 5.5631581)

 - Block number: 2 starting at 2015-11-09T04:41:16.348.096.750 (XO: 29.3482387, t=35.6°C)  Rank: 00003
   TM values: Max =     4  Min =   -12  Ave =    -4.62  Std =     2.05
   Min TM value:-12 at an offset of 8.325 usec -> 2015-11-09T04:41:16.348.105.075 (XO: 29.3482470)
   Max TM value:4 at an offset of 69.700 usec -> 2015-11-09T04:41:16.348.166.450 (XO: 29.3483084)
   Max dTM/dt [1/us]::6.40e+01 at an offset of 179.863 usec -> 2015-11-09T04:41:16.348.276.613 (XO: 29.3484186)
   Min dTM/dt [1/us]::-7.10e+01 at an offset of 150.913 usec -> 2015-11-09T04:41:16.348.247.662 (XO: 29.3483896)

 - Block number: 3 starting at 2015-11-09T04:41:17.022.882.750 (XO: 30.0230247, t=35.6°C)  Rank: 00002
   TM values: Max =     4  Min =   -11  Ave =    -4.58  Std =     1.94
   Min TM value:-11 at an offset of 0.013 usec -> 2015-11-09T04:41:17.022.882.762 (XO: 30.0230247)
   Max TM value:4 at an offset of 96.825 usec -> 2015-11-09T04:41:17.022.979.575 (XO: 30.0231215)
   Max dTM/dt [1/us]::6.60e+01 at an offset of 120.888 usec -> 2015-11-09T04:41:17.023.003.637 (XO: 30.0231456)
   Min dTM/dt [1/us]::-7.20e+01 at an offset of 228.088 usec -> 2015-11-09T04:41:17.023.110.838 (XO: 30.0232528)
