-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151101_220002_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.594163 to 39.965109
-each separately from 6s to 39s and from 461412us to 965109us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.866960 in 1 packets - #/No: 18/18
      16.503119 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-01T22:00:11.594.163.000 to 2015-11-01T22:00:44.965.109.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.866960 in 1 packets - #/No: 18/18
      16.503119 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.5328667 to 40.5327712
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999509 in 1 packets - #/No: 18/18
      15.9999536 in 1 packets - #/No: 36/18
Temperature FPGA: 
      08 -> 44.6°C in 54 packets (all packets)
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151101_220002_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.1270294 to 8.1273184
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.8669110  -- OK for SL --
   MIU time from 2015-11-01T22:00:11.594.163.000 to 2015-11-01T22:00:11.594.452.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.866960

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.9942294 to 24.9945184
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5030710  -- OK for SL --
   MIU time from 2015-11-01T22:00:28.461.412.000 to 2015-11-01T22:00:28.461.701.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.503119

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.4975894 to 41.4978784
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-01T22:00:44.964.820.000 to 2015-11-01T22:00:44.965.109.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-01T22:00:11.594.022.750
   Max SL corrected packet time: 2015-11-01T22:00:44.964.968.750
   Min SL corrected XO rel time: 6.5941644
   Max SL corrected XO rel time: 39.9651100
   TM values from -35 to 26 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-01T22:00:11.594.022.750 (XO: 6.5941644, t=32.4°C)  Rank: 00007
   TM values: Max =    23  Min =   -31  Ave =    -4.74  Std =     7.25
   Min TM value:-31 at an offset of 127.050 usec -> 2015-11-01T22:00:11.594.149.800 (XO: 6.5942915)
   Max TM value:23 at an offset of 126.313 usec -> 2015-11-01T22:00:11.594.149.062 (XO: 6.5942907)
   Max dTM/dt [1/us]::3.81e+02 at an offset of 126.125 usec -> 2015-11-01T22:00:11.594.148.875 (XO: 6.5942905)
   Min dTM/dt [1/us]::-3.80e+02 at an offset of 127.013 usec -> 2015-11-01T22:00:11.594.149.762 (XO: 6.5942914)

 - Block number: 2 starting at 2015-11-01T22:00:28.461.271.750 (XO: 23.4614131, t=32.4°C)  Rank: 00008
   TM values: Max =    26  Min =   -35  Ave =    -4.70  Std =     8.77
   Min TM value:-35 at an offset of 62.775 usec -> 2015-11-01T22:00:28.461.334.525 (XO: 23.4614759)
   Max TM value:26 at an offset of 63.350 usec -> 2015-11-01T22:00:28.461.335.100 (XO: 23.4614765)
   Max dTM/dt [1/us]::5.09e+02 at an offset of 56.825 usec -> 2015-11-01T22:00:28.461.328.575 (XO: 23.4614700)
   Min dTM/dt [1/us]::-4.90e+02 at an offset of 56.750 usec -> 2015-11-01T22:00:28.461.328.500 (XO: 23.4614699)

 - Block number: 3 starting at 2015-11-01T22:00:44.964.679.750 (XO: 39.9648210, t=32.4°C)  Rank: 00010
   TM values: Max =    22  Min =   -32  Ave =    -4.74  Std =     7.52
   Min TM value:-32 at an offset of 127.788 usec -> 2015-11-01T22:00:44.964.807.537 (XO: 39.9649488)
   Max TM value:22 at an offset of 139.700 usec -> 2015-11-01T22:00:44.964.819.450 (XO: 39.9649607)
   Max dTM/dt [1/us]::4.60e+02 at an offset of 128.638 usec -> 2015-11-01T22:00:44.964.808.387 (XO: 39.9649496)
   Min dTM/dt [1/us]::-4.59e+02 at an offset of 129.213 usec -> 2015-11-01T22:00:44.964.808.962 (XO: 39.9649502)
