-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151031_135150_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.003740 to 44.828670
-each separately from 0s to 44s and from 3740us to 840627us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.836309 in 1 packets - #/No: 18/18
      25.987754 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-31T13:51:53.003.740.000 to 2015-10-31T13:52:37.828.670.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.836309 in 1 packets - #/No: 18/18
      25.987754 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.3905442 to 45.3904173
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999481 in 1 packets - #/No: 18/18
      25.9999250 in 1 packets - #/No: 36/18
Temperature FPGA: 
      08 -> 44.6°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151031_135150_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.3942849 to 1.3945739
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8362550  -- OK for SL --
   MIU time from 2015-10-31T13:51:53.003.740.000 to 2015-10-31T13:51:53.004.029.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.836309

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.2308289 to 20.2311179
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.9876790  -- OK for SL --
   MIU time from 2015-10-31T13:52:11.840.338.000 to 2015-10-31T13:52:11.840.627.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.987754

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.2187969 to 46.2190859
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-31T13:52:37.828.381.000 to 2015-10-31T13:52:37.828.670.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-31T13:51:53.003.599.750
   Max SL corrected packet time: 2015-10-31T13:52:37.828.529.750
   Min SL corrected XO rel time: 0.0037407
   Max SL corrected XO rel time: 44.8286717
   TM values from -27 to 16 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-31T13:51:53.003.599.750 (XO: 0.0037407, t=32.4°C)  Rank: 00002
   TM values: Max =     8  Min =   -16  Ave =    -4.82  Std =     3.44
   Min TM value:-16 at an offset of 117.825 usec -> 2015-10-31T13:51:53.003.717.575 (XO: 0.0038585)
   Max TM value:8 at an offset of 290.301 usec -> 2015-10-31T13:51:53.003.890.050 (XO: 0.0040310)
   Max dTM/dt [1/us]::1.10e+02 at an offset of 81.613 usec -> 2015-10-31T13:51:53.003.681.362 (XO: 0.0038223)
   Min dTM/dt [1/us]::-1.04e+02 at an offset of 7.600 usec -> 2015-10-31T13:51:53.003.607.350 (XO: 0.0037483)

 - Block number: 2 starting at 2015-10-31T13:52:11.840.197.750 (XO: 18.8403390, t=32.4°C)  Rank: 00003
   TM values: Max =    14  Min =   -23  Ave =    -4.71  Std =     5.26
   Min TM value:-23 at an offset of 67.063 usec -> 2015-10-31T13:52:11.840.264.812 (XO: 18.8404060)
   Max TM value:14 at an offset of 67.925 usec -> 2015-10-31T13:52:11.840.265.675 (XO: 18.8404069)
   Max dTM/dt [1/us]::1.29e+02 at an offset of 305.988 usec -> 2015-10-31T13:52:11.840.503.738 (XO: 18.8406449)
   Min dTM/dt [1/us]::-1.10e+02 at an offset of 77.250 usec -> 2015-10-31T13:52:11.840.275.000 (XO: 18.8404162)

 - Block number: 3 starting at 2015-10-31T13:52:37.828.240.750 (XO: 44.8283827, t=32.4°C)  Rank: 00005
   TM values: Max =    16  Min =   -27  Ave =    -4.71  Std =     7.07
   Min TM value:-27 at an offset of 174.876 usec -> 2015-10-31T13:52:37.828.415.625 (XO: 44.8285576)
   Max TM value:16 at an offset of 137.688 usec -> 2015-10-31T13:52:37.828.378.437 (XO: 44.8285204)
   Max dTM/dt [1/us]::1.00e+02 at an offset of 50.538 usec -> 2015-10-31T13:52:37.828.291.287 (XO: 44.8284332)
   Min dTM/dt [1/us]::-1.32e+02 at an offset of 0.013 usec -> 2015-10-31T13:52:37.828.240.762 (XO: 44.8283827)
