-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151030_201426_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.457052 to 37.344007
-each separately from 11s to 37s and from 343718us to 707718us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.250088 in 1 packets - #/No: 18/18
      11.636000 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-30T20:14:40.457.052.000 to 2015-10-30T20:15:06.344.007.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.250088 in 1 packets - #/No: 18/18
      11.636000 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.5038497 to 38.5037748
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999597 in 1 packets - #/No: 18/18
      11.9999654 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      08 -> 44.6°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151030_201426_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.9609010 to 12.9611900
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2500470  -- OK for SL --
   MIU time from 2015-10-30T20:14:40.457.052.000 to 2015-10-30T20:14:40.457.341.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.250088

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.2112370 to 27.2115260
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.6359670  -- OK for SL --
   MIU time from 2015-10-30T20:14:54.707.429.000 to 2015-10-30T20:14:54.707.718.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.636000

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.8474930 to 38.8477820
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-30T20:15:06.343.718.000 to 2015-10-30T20:15:06.344.007.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-30T20:14:40.456.911.750
   Max SL corrected packet time: 2015-10-30T20:15:06.343.866.750
   Min SL corrected XO rel time: 11.4570526
   Max SL corrected XO rel time: 37.3440082
   TM values from -31 to 21 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-30T20:14:40.456.911.750 (XO: 11.4570526, t=32.4°C)  Rank: 00005
   TM values: Max =    15  Min =   -25  Ave =    -4.63  Std =     5.37
   Min TM value:-25 at an offset of 112.725 usec -> 2015-10-30T20:14:40.457.024.475 (XO: 11.4571654)
   Max TM value:15 at an offset of 113.488 usec -> 2015-10-30T20:14:40.457.025.237 (XO: 11.4571661)
   Max dTM/dt [1/us]::2.92e+02 at an offset of 106.588 usec -> 2015-10-30T20:14:40.457.018.337 (XO: 11.4571592)
   Min dTM/dt [1/us]::-2.93e+02 at an offset of 112.688 usec -> 2015-10-30T20:14:40.457.024.437 (XO: 11.4571653)

 - Block number: 2 starting at 2015-10-30T20:14:54.707.288.750 (XO: 25.7074297, t=32.4°C)  Rank: 00005
   TM values: Max =    21  Min =   -31  Ave =    -4.77  Std =     6.32
   Min TM value:-31 at an offset of 126.700 usec -> 2015-10-30T20:14:54.707.415.450 (XO: 25.7075564)
   Max TM value:21 at an offset of 87.525 usec -> 2015-10-30T20:14:54.707.376.275 (XO: 25.7075172)
   Max dTM/dt [1/us]::3.67e+02 at an offset of 116.088 usec -> 2015-10-30T20:14:54.707.404.837 (XO: 25.7075457)
   Min dTM/dt [1/us]::-3.62e+02 at an offset of 116.150 usec -> 2015-10-30T20:14:54.707.404.900 (XO: 25.7075458)

 - Block number: 3 starting at 2015-10-30T20:15:06.343.577.750 (XO: 37.3437192, t=32.4°C)  Rank: 00005
   TM values: Max =    18  Min =   -24  Ave =    -4.74  Std =     5.33
   Min TM value:-24 at an offset of 65.538 usec -> 2015-10-30T20:15:06.343.643.287 (XO: 37.3437847)
   Max TM value:18 at an offset of 106.088 usec -> 2015-10-30T20:15:06.343.683.837 (XO: 37.3438253)
   Max dTM/dt [1/us]::2.93e+02 at an offset of 109.538 usec -> 2015-10-30T20:15:06.343.687.287 (XO: 37.3438287)
   Min dTM/dt [1/us]::-3.10e+02 at an offset of 109.463 usec -> 2015-10-30T20:15:06.343.687.212 (XO: 37.3438286)
