-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151028_214538_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.741942 to 37.567308
-each separately from 7s to 37s and from 259985us to 742231us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.517754 in 1 packets - #/No: 18/18
      9.306745 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-28T21:45:48.741.942.000 to 2015-10-28T21:46:18.567.308.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.517754 in 1 packets - #/No: 18/18
      9.306745 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4545042 to 38.4544186
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999401 in 1 packets - #/No: 18/18
      8.9999743 in 1 packets - #/No: 36/18
Temperature FPGA: 
      09 -> 43.3°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      07 -> 45.9°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151028_214538_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.1964452 to 9.1967342
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5176950  -- OK for SL --
   MIU time from 2015-10-28T21:45:48.741.942.000 to 2015-10-28T21:45:48.742.231.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.517754

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.7144292 to 29.7147182
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.3067190  -- OK for SL --
   MIU time from 2015-10-28T21:46:09.259.985.000 to 2015-10-28T21:46:09.260.274.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.306745

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.0214372 to 39.0217262
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-28T21:46:18.567.019.000 to 2015-10-28T21:46:18.567.308.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-28T21:45:48.741.801.750
   Max SL corrected packet time: 2015-10-28T21:46:18.567.167.750
   Min SL corrected XO rel time: 7.7419432
   Max SL corrected XO rel time: 37.5673092
   TM values from -53 to 48 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-28T21:45:48.741.801.750 (XO: 7.7419432, t=34.0°C)  Rank: 00015
   TM values: Max =    22  Min =   -34  Ave =    -4.77  Std =     5.82
   Min TM value:-34 at an offset of 74.025 usec -> 2015-10-28T21:45:48.741.875.775 (XO: 7.7420172)
   Max TM value:22 at an offset of 118.450 usec -> 2015-10-28T21:45:48.741.920.200 (XO: 7.7420616)
   Max dTM/dt [1/us]::2.46e+02 at an offset of 270.001 usec -> 2015-10-28T21:45:48.742.071.750 (XO: 7.7422132)
   Min dTM/dt [1/us]::-2.69e+02 at an offset of 183.376 usec -> 2015-10-28T21:45:48.741.985.125 (XO: 7.7421265)

 - Block number: 2 starting at 2015-10-28T21:46:09.259.844.750 (XO: 28.2599857, t=34.0°C)  Rank: 00013
   TM values: Max =    48  Min =   -53  Ave =    -4.76  Std =     9.46
   Min TM value:-53 at an offset of 132.913 usec -> 2015-10-28T21:46:09.259.977.662 (XO: 28.2601186)
   Max TM value:48 at an offset of 132.575 usec -> 2015-10-28T21:46:09.259.977.325 (XO: 28.2601182)
   Max dTM/dt [1/us]::4.43e+02 at an offset of 133.650 usec -> 2015-10-28T21:46:09.259.978.400 (XO: 28.2601193)
   Min dTM/dt [1/us]::-5.35e+02 at an offset of 132.738 usec -> 2015-10-28T21:46:09.259.977.487 (XO: 28.2601184)

 - Block number: 3 starting at 2015-10-28T21:46:18.566.878.750 (XO: 37.5670202, t=34.0°C)  Rank: 00015
   TM values: Max =    35  Min =   -39  Ave =    -4.82  Std =     7.35
   Min TM value:-39 at an offset of 80.925 usec -> 2015-10-28T21:46:18.566.959.675 (XO: 37.5671011)
   Max TM value:35 at an offset of 79.025 usec -> 2015-10-28T21:46:18.566.957.775 (XO: 37.5670992)
   Max dTM/dt [1/us]::5.07e+02 at an offset of 193.176 usec -> 2015-10-28T21:46:18.567.071.925 (XO: 37.5672134)
   Min dTM/dt [1/us]::-3.38e+02 at an offset of 227.776 usec -> 2015-10-28T21:46:18.567.106.525 (XO: 37.5672480)
