-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151027_154124_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.696671 to 30.146005
-each separately from 0s to 30s and from 145716us to 983529us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      25.286280 in 1 packets - #/No: 18/18
      4.162187 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-27T15:41:27.696.671.000 to 2015-10-27T15:41:57.146.005.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      25.286280 in 1 packets - #/No: 18/18
      4.162187 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4658077 to 31.4657216
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.9999283 in 1 packets - #/No: 18/18
      4.9999856 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151027_154124_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1624779 to 2.1627669
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.2862070  -- OK for SL --
   MIU time from 2015-10-27T15:41:27.696.671.000 to 2015-10-27T15:41:27.696.960.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.286280

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.4489739 to 27.4492629
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.1621750  -- OK for SL --
   MIU time from 2015-10-27T15:41:52.983.240.000 to 2015-10-27T15:41:52.983.529.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.162187

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.6114379 to 31.6117269
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-27T15:41:57.145.716.000 to 2015-10-27T15:41:57.146.005.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-27T15:41:27.696.530.750
   Max SL corrected packet time: 2015-10-27T15:41:57.145.864.750
   Min SL corrected XO rel time: 0.6966722
   Max SL corrected XO rel time: 30.1460057
   TM values from -29 to 25 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-27T15:41:27.696.530.750 (XO: 0.6966722, t=34.0°C)  Rank: 00003
   TM values: Max =    20  Min =   -27  Ave =    -4.70  Std =     6.20
   Min TM value:-27 at an offset of 21.975 usec -> 2015-10-27T15:41:27.696.552.725 (XO: 0.6966942)
   Max TM value:20 at an offset of 222.338 usec -> 2015-10-27T15:41:27.696.753.088 (XO: 0.6968946)
   Max dTM/dt [1/us]::2.04e+02 at an offset of 93.038 usec -> 2015-10-27T15:41:27.696.623.787 (XO: 0.6967653)
   Min dTM/dt [1/us]::-2.28e+02 at an offset of 235.013 usec -> 2015-10-27T15:41:27.696.765.763 (XO: 0.6969072)

 - Block number: 2 starting at 2015-10-27T15:41:52.983.099.750 (XO: 25.9832408, t=34.0°C)  Rank: 00005
   TM values: Max =    25  Min =   -29  Ave =    -4.70  Std =     6.77
   Min TM value:-29 at an offset of 53.363 usec -> 2015-10-27T15:41:52.983.153.112 (XO: 25.9832941)
   Max TM value:25 at an offset of 44.125 usec -> 2015-10-27T15:41:52.983.143.875 (XO: 25.9832849)
   Max dTM/dt [1/us]::3.13e+02 at an offset of 201.451 usec -> 2015-10-27T15:41:52.983.301.200 (XO: 25.9834422)
   Min dTM/dt [1/us]::-2.76e+02 at an offset of 217.088 usec -> 2015-10-27T15:41:52.983.316.838 (XO: 25.9834578)

 - Block number: 3 starting at 2015-10-27T15:41:57.145.575.750 (XO: 30.1457167, t=34.0°C)  Rank: 00003
   TM values: Max =    15  Min =   -26  Ave =    -4.71  Std =     5.27
   Min TM value:-26 at an offset of 120.863 usec -> 2015-10-27T15:41:57.145.696.612 (XO: 30.1458376)
   Max TM value:15 at an offset of 119.975 usec -> 2015-10-27T15:41:57.145.695.725 (XO: 30.1458367)
   Max dTM/dt [1/us]::1.93e+02 at an offset of 262.613 usec -> 2015-10-27T15:41:57.145.838.363 (XO: 30.1459793)
   Min dTM/dt [1/us]::-2.05e+02 at an offset of 162.913 usec -> 2015-10-27T15:41:57.145.738.662 (XO: 30.1458796)
