-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151026_223505_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.000953 to 38.173543
-each separately from 0s to 38s and from 953us to 986178us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.984647 in 1 packets - #/No: 18/18
      13.187076 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-26T22:35:08.000.953.000 to 2015-10-26T22:35:46.173.543.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.984647 in 1 packets - #/No: 18/18
      13.187076 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4132183 to 39.4131102
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999317 in 1 packets - #/No: 18/18
      13.9999602 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 54 packets (all packets)
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151026_223505_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4141728 to 1.4144618
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.9845750  -- OK for SL --
   MIU time from 2015-10-26T22:35:08.000.953.000 to 2015-10-26T22:35:08.001.242.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.984647

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.3990368 to 26.3993258
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1870390  -- OK for SL --
   MIU time from 2015-10-26T22:35:32.985.889.000 to 2015-10-26T22:35:32.986.178.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.187076

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.5863648 to 39.5866538
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-26T22:35:46.173.254.000 to 2015-10-26T22:35:46.173.543.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-26T22:35:08.000.812.750
   Max SL corrected packet time: 2015-10-26T22:35:46.173.402.750
   Min SL corrected XO rel time: 0.0009545
   Max SL corrected XO rel time: 38.1735441
   TM values from -20 to 11 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-26T22:35:08.000.812.750 (XO: 0.0009545, t=32.4°C)  Rank: 00002
   TM values: Max =     7  Min =   -18  Ave =    -4.70  Std =     3.52
   Min TM value:-18 at an offset of 37.375 usec -> 2015-10-26T22:35:08.000.850.125 (XO: 0.0009919)
   Max TM value:7 at an offset of 20.525 usec -> 2015-10-26T22:35:08.000.833.275 (XO: 0.0009750)
   Max dTM/dt [1/us]::1.38e+02 at an offset of 141.350 usec -> 2015-10-26T22:35:08.000.954.100 (XO: 0.0010959)
   Min dTM/dt [1/us]::-1.61e+02 at an offset of 44.625 usec -> 2015-10-26T22:35:08.000.857.375 (XO: 0.0009991)

 - Block number: 2 starting at 2015-10-26T22:35:32.985.748.750 (XO: 24.9858896, t=32.4°C)  Rank: 00003
   TM values: Max =    11  Min =   -20  Ave =    -4.64  Std =     4.34
   Min TM value:-20 at an offset of 208.713 usec -> 2015-10-26T22:35:32.985.957.463 (XO: 24.9860983)
   Max TM value:11 at an offset of 46.525 usec -> 2015-10-26T22:35:32.985.795.275 (XO: 24.9859361)
   Max dTM/dt [1/us]::1.87e+02 at an offset of 97.738 usec -> 2015-10-26T22:35:32.985.846.487 (XO: 24.9859873)
   Min dTM/dt [1/us]::-1.91e+02 at an offset of 276.638 usec -> 2015-10-26T22:35:32.986.025.388 (XO: 24.9861662)

 - Block number: 3 starting at 2015-10-26T22:35:46.173.113.750 (XO: 38.1732551, t=32.4°C)  Rank: 00003
   TM values: Max =     9  Min =   -18  Ave =    -4.65  Std =     3.73
   Min TM value:-18 at an offset of 98.838 usec -> 2015-10-26T22:35:46.173.212.587 (XO: 38.1733539)
   Max TM value:9 at an offset of 58.525 usec -> 2015-10-26T22:35:46.173.172.275 (XO: 38.1733136)
   Max dTM/dt [1/us]::1.42e+02 at an offset of 100.650 usec -> 2015-10-26T22:35:46.173.214.400 (XO: 38.1733557)
   Min dTM/dt [1/us]::-1.39e+02 at an offset of 118.513 usec -> 2015-10-26T22:35:46.173.232.262 (XO: 38.1733736)
