-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151023_212959_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.499852 to 32.061785
-each separately from 5s to 32s and from 61496us to 500141us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.798660 in 1 packets - #/No: 18/18
      13.762406 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-23T21:30:07.499.852.000 to 2015-10-23T21:30:34.061.785.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.798660 in 1 packets - #/No: 18/18
      13.762406 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4374573 to 33.4373802
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.9999629 in 1 packets - #/No: 18/18
      13.9999600 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151023_212959_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.9373094 to 6.9375984
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.7986230  -- OK for SL --
   MIU time from 2015-10-23T21:30:07.499.852.000 to 2015-10-23T21:30:07.500.141.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.798660

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.7362214 to 19.7365104
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.7623670  -- OK for SL --
   MIU time from 2015-10-23T21:30:20.298.801.000 to 2015-10-23T21:30:20.299.090.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.762406

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.4988774 to 33.4991664
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-23T21:30:34.061.496.000 to 2015-10-23T21:30:34.061.785.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-23T21:30:07.499.711.750
   Max SL corrected packet time: 2015-10-23T21:30:34.061.644.750
   Min SL corrected XO rel time: 5.4998535
   Max SL corrected XO rel time: 32.0617864
   TM values from -29 to 21 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-23T21:30:07.499.711.750 (XO: 5.4998535, t=34.0°C)  Rank: 00004
   TM values: Max =    20  Min =   -28  Ave =    -4.71  Std =     6.29
   Min TM value:-28 at an offset of 271.526 usec -> 2015-10-23T21:30:07.499.983.275 (XO: 5.5001250)
   Max TM value:20 at an offset of 91.988 usec -> 2015-10-23T21:30:07.499.803.737 (XO: 5.4999455)
   Max dTM/dt [1/us]::3.17e+02 at an offset of 121.950 usec -> 2015-10-23T21:30:07.499.833.700 (XO: 5.4999754)
   Min dTM/dt [1/us]::-2.98e+02 at an offset of 271.488 usec -> 2015-10-23T21:30:07.499.983.238 (XO: 5.5001250)

 - Block number: 2 starting at 2015-10-23T21:30:20.298.660.750 (XO: 18.2988021, t=34.0°C)  Rank: 00004
   TM values: Max =    21  Min =   -29  Ave =    -4.67  Std =     6.48
   Min TM value:-29 at an offset of 123.075 usec -> 2015-10-23T21:30:20.298.783.825 (XO: 18.2989251)
   Max TM value:21 at an offset of 123.538 usec -> 2015-10-23T21:30:20.298.784.287 (XO: 18.2989256)
   Max dTM/dt [1/us]::3.15e+02 at an offset of 124.463 usec -> 2015-10-23T21:30:20.298.785.212 (XO: 18.2989265)
   Min dTM/dt [1/us]::-3.28e+02 at an offset of 124.400 usec -> 2015-10-23T21:30:20.298.785.150 (XO: 18.2989265)

 - Block number: 3 starting at 2015-10-23T21:30:34.061.355.750 (XO: 32.0614974, t=34.0°C)  Rank: 00004
   TM values: Max =    18  Min =   -27  Ave =    -4.71  Std =     6.36
   Min TM value:-27 at an offset of 234.363 usec -> 2015-10-23T21:30:34.061.590.113 (XO: 32.0617317)
   Max TM value:18 at an offset of 103.888 usec -> 2015-10-23T21:30:34.061.459.637 (XO: 32.0616013)
   Max dTM/dt [1/us]::3.36e+02 at an offset of 234.688 usec -> 2015-10-23T21:30:34.061.590.438 (XO: 32.0617321)
   Min dTM/dt [1/us]::-3.07e+02 at an offset of 234.338 usec -> 2015-10-23T21:30:34.061.590.088 (XO: 32.0617317)
