-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151021_214847_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 14.670759 to 32.572506
-each separately from 14s to 32s and from 572217us to 671048us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.951361 in 1 packets - #/No: 18/18
      16.949519 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-21T21:49:04.670.759.000 to 2015-10-21T21:49:22.572.506.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.951361 in 1 packets - #/No: 18/18
      16.949519 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4858623 to 33.4858113
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.9999972 in 1 packets - #/No: 18/18
      16.9999518 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      08 -> 44.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151021_214847_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.1566200 to 16.1569090
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.9513590  -- OK for SL --
   MIU time from 2015-10-21T21:49:04.670.759.000 to 2015-10-21T21:49:04.671.048.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.951361

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1082680 to 17.1085570
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9494710  -- OK for SL --
   MIU time from 2015-10-21T21:49:05.622.409.000 to 2015-10-21T21:49:05.622.698.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.949519

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.0580280 to 34.0583170
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-21T21:49:22.572.217.000 to 2015-10-21T21:49:22.572.506.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-21T21:49:04.670.618.750
   Max SL corrected packet time: 2015-10-21T21:49:22.572.365.750
   Min SL corrected XO rel time: 14.6707596
   Max SL corrected XO rel time: 32.5725074
   TM values from -31 to 28 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-21T21:49:04.670.618.750 (XO: 14.6707596, t=32.4°C)  Rank: 00010
   TM values: Max =    28  Min =   -31  Ave =    -4.82  Std =     6.50
   Min TM value:-31 at an offset of 171.075 usec -> 2015-10-21T21:49:04.670.789.825 (XO: 14.6709307)
   Max TM value:28 at an offset of 141.975 usec -> 2015-10-21T21:49:04.670.760.725 (XO: 14.6709016)
   Max dTM/dt [1/us]::3.01e+02 at an offset of 141.775 usec -> 2015-10-21T21:49:04.670.760.525 (XO: 14.6709014)
   Min dTM/dt [1/us]::-2.70e+02 at an offset of 147.738 usec -> 2015-10-21T21:49:04.670.766.487 (XO: 14.6709073)

 - Block number: 2 starting at 2015-10-21T21:49:05.622.268.750 (XO: 15.6224103, t=32.4°C)  Rank: 00006
   TM values: Max =    28  Min =   -27  Ave =    -4.82  Std =     5.64
   Min TM value:-27 at an offset of 81.413 usec -> 2015-10-21T21:49:05.622.350.162 (XO: 15.6224917)
   Max TM value:28 at an offset of 78.513 usec -> 2015-10-21T21:49:05.622.347.262 (XO: 15.6224888)
   Max dTM/dt [1/us]::2.60e+02 at an offset of 187.701 usec -> 2015-10-21T21:49:05.622.456.450 (XO: 15.6225980)
   Min dTM/dt [1/us]::-2.55e+02 at an offset of 186.951 usec -> 2015-10-21T21:49:05.622.455.700 (XO: 15.6225973)

 - Block number: 3 starting at 2015-10-21T21:49:22.572.076.750 (XO: 32.5722184, t=32.4°C)  Rank: 00004
   TM values: Max =    26  Min =   -27  Ave =    -4.76  Std =     4.42
   Min TM value:-27 at an offset of 221.551 usec -> 2015-10-21T21:49:22.572.298.300 (XO: 32.5724399)
   Max TM value:26 at an offset of 117.900 usec -> 2015-10-21T21:49:22.572.194.650 (XO: 32.5723363)
   Max dTM/dt [1/us]::2.16e+02 at an offset of 222.313 usec -> 2015-10-21T21:49:22.572.299.063 (XO: 32.5724407)
   Min dTM/dt [1/us]::-2.19e+02 at an offset of 222.076 usec -> 2015-10-21T21:49:22.572.298.825 (XO: 32.5724404)
