-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151017_041748_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.010362 to 41.593741
-each separately from 0s to 41s and from 10362us to 593741us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.023105 in 1 packets - #/No: 18/18
      17.559407 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-17T04:17:51.010.362.000 to 2015-10-17T04:18:32.593.741.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.023105 in 1 packets - #/No: 18/18
      17.559407 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4300143 to 42.4299014
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999340 in 1 packets - #/No: 18/18
      16.9999531 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      C0 -> 35.6°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151017_041748_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4403772 to 1.4406662
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.0230390  -- OK for SL --
   MIU time from 2015-10-17T04:17:51.010.362.000 to 2015-10-17T04:17:51.010.651.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.023105

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.4637052 to 25.4639942
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5593590  -- OK for SL --
   MIU time from 2015-10-17T04:18:15.033.756.000 to 2015-10-17T04:18:15.034.045.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.559407

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.0233532 to 43.0236422
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-17T04:18:32.593.452.000 to 2015-10-17T04:18:32.593.741.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-17T04:17:51.010.221.750
   Max SL corrected packet time: 2015-10-17T04:18:32.593.600.750
   Min SL corrected XO rel time: 0.0103629
   Max SL corrected XO rel time: 41.5937424
   TM values from -18 to 8 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-17T04:17:51.010.221.750 (XO: 0.0103629, t=37.2°C)  Rank: 00002
   TM values: Max =     5  Min =   -15  Ave =    -4.68  Std =     2.60
   Min TM value:-15 at an offset of 178.000 usec -> 2015-10-17T04:17:51.010.399.750 (XO: 0.0105409)
   Max TM value:5 at an offset of 186.551 usec -> 2015-10-17T04:17:51.010.408.300 (XO: 0.0105495)
   Max dTM/dt [1/us]::9.80e+01 at an offset of 9.375 usec -> 2015-10-17T04:17:51.010.231.125 (XO: 0.0103723)
   Min dTM/dt [1/us]::-8.10e+01 at an offset of 177.963 usec -> 2015-10-17T04:17:51.010.399.712 (XO: 0.0105409)

 - Block number: 2 starting at 2015-10-17T04:18:15.033.615.750 (XO: 24.0337570, t=37.2°C)  Rank: 00003
   TM values: Max =     8  Min =   -17  Ave =    -4.82  Std =     3.37
   Min TM value:-17 at an offset of 123.325 usec -> 2015-10-17T04:18:15.033.739.075 (XO: 24.0338803)
   Max TM value:8 at an offset of 123.913 usec -> 2015-10-17T04:18:15.033.739.662 (XO: 24.0338809)
   Max dTM/dt [1/us]::9.40e+01 at an offset of 172.888 usec -> 2015-10-17T04:18:15.033.788.637 (XO: 24.0339299)
   Min dTM/dt [1/us]::-1.03e+02 at an offset of 205.276 usec -> 2015-10-17T04:18:15.033.821.025 (XO: 24.0339623)

 - Block number: 3 starting at 2015-10-17T04:18:32.593.311.750 (XO: 41.5934534, t=37.2°C)  Rank: 00003
   TM values: Max =     8  Min =   -18  Ave =    -4.76  Std =     3.57
   Min TM value:-18 at an offset of 86.100 usec -> 2015-10-17T04:18:32.593.397.850 (XO: 41.5935395)
   Max TM value:8 at an offset of 77.663 usec -> 2015-10-17T04:18:32.593.389.412 (XO: 41.5935311)
   Max dTM/dt [1/us]::1.13e+02 at an offset of 282.613 usec -> 2015-10-17T04:18:32.593.594.363 (XO: 41.5937360)
   Min dTM/dt [1/us]::-1.28e+02 at an offset of 117.713 usec -> 2015-10-17T04:18:32.593.429.462 (XO: 41.5935711)
