-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151015_144715_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.248332 to 34.511511
-each separately from 8s to 34s and from 248332us to 913055us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.664145 in 1 packets - #/No: 18/18
      8.598167 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-15T14:47:26.248.332.000 to 2015-10-15T14:47:52.511.511.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.664145 in 1 packets - #/No: 18/18
      8.598167 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4212033 to 35.4211301
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999521 in 1 packets - #/No: 18/18
      8.9999747 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151015_144715_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.6695361 to 9.6698251
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6640950  -- OK for SL --
   MIU time from 2015-10-15T14:47:26.248.332.000 to 2015-10-15T14:47:26.248.621.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.664145

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.3339201 to 27.3342091
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.5981430  -- OK for SL --
   MIU time from 2015-10-15T14:47:43.912.766.000 to 2015-10-15T14:47:43.913.055.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.598167

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.9323521 to 35.9326411
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-15T14:47:52.511.222.000 to 2015-10-15T14:47:52.511.511.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-15T14:47:26.248.191.750
   Max SL corrected packet time: 2015-10-15T14:47:52.511.370.750
   Min SL corrected XO rel time: 8.2483335
   Max SL corrected XO rel time: 34.5115124
   TM values from -50 to 36 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-15T14:47:26.248.191.750 (XO: 8.2483335, t=34.0°C)  Rank: 00015
   TM values: Max =    36  Min =   -50  Ave =    -4.75  Std =     6.95
   Min TM value:-50 at an offset of 131.650 usec -> 2015-10-15T14:47:26.248.323.400 (XO: 8.2484651)
   Max TM value:36 at an offset of 146.350 usec -> 2015-10-15T14:47:26.248.338.100 (XO: 8.2484798)
   Max dTM/dt [1/us]::2.10e+02 at an offset of 110.800 usec -> 2015-10-15T14:47:26.248.302.550 (XO: 8.2484443)
   Min dTM/dt [1/us]::-2.00e+02 at an offset of 110.888 usec -> 2015-10-15T14:47:26.248.302.637 (XO: 8.2484444)

 - Block number: 2 starting at 2015-10-15T14:47:43.912.625.750 (XO: 25.9127673, t=34.0°C)  Rank: 00003
   TM values: Max =    18  Min =   -22  Ave =    -4.65  Std =     5.18
   Min TM value:-22 at an offset of 25.538 usec -> 2015-10-15T14:47:43.912.651.287 (XO: 25.9127928)
   Max TM value:18 at an offset of 93.188 usec -> 2015-10-15T14:47:43.912.718.937 (XO: 25.9128604)
   Max dTM/dt [1/us]::1.09e+02 at an offset of 267.951 usec -> 2015-10-15T14:47:43.912.893.700 (XO: 25.9130352)
   Min dTM/dt [1/us]::-1.15e+02 at an offset of 0.000 usec -> 2015-10-15T14:47:43.912.625.750 (XO: 25.9127673)

 - Block number: 3 starting at 2015-10-15T14:47:52.511.081.750 (XO: 34.5112234, t=34.0°C)  Rank: 00003
   TM values: Max =    10  Min =   -21  Ave =    -4.60  Std =     4.18
   Min TM value:-21 at an offset of 225.488 usec -> 2015-10-15T14:47:52.511.307.238 (XO: 34.5114489)
   Max TM value:10 at an offset of 142.325 usec -> 2015-10-15T14:47:52.511.224.075 (XO: 34.5113658)
   Max dTM/dt [1/us]::1.12e+02 at an offset of 304.276 usec -> 2015-10-15T14:47:52.511.386.025 (XO: 34.5115277)
   Min dTM/dt [1/us]::-1.11e+02 at an offset of 33.313 usec -> 2015-10-15T14:47:52.511.115.062 (XO: 34.5112567)
