-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151013_181239_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.334621 to 31.278617
-each separately from 10s to 31s and from 164079us to 334910us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.829169 in 1 packets - #/No: 18/18
      3.113960 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-13T18:12:52.334.621.000 to 2015-10-13T18:13:13.278.617.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.829169 in 1 packets - #/No: 18/18
      3.113960 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4566444 to 32.4565853
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999494 in 1 packets - #/No: 18/18
      2.9999915 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      BF -> 34.0°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151013_181239_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.7912659 to 11.7915549
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8291190  -- OK for SL --
   MIU time from 2015-10-13T18:12:52.334.621.000 to 2015-10-13T18:12:52.334.910.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.829169

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      34 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.6206739 to 29.6209629
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.1139510  -- OK for SL --
   MIU time from 2015-10-13T18:13:10.164.079.000 to 2015-10-13T18:13:10.164.368.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.113960

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.7349139 to 32.7352029
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-13T18:13:13.278.328.000 to 2015-10-13T18:13:13.278.617.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-13T18:12:52.334.480.750
   Max SL corrected packet time: 2015-10-13T18:13:13.278.476.750
   Min SL corrected XO rel time: 10.3346224
   Max SL corrected XO rel time: 31.2786184
   TM values from -45 to 37 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-13T18:12:52.334.480.750 (XO: 10.3346224, t=35.6°C)  Rank: 00007
   TM values: Max =    28  Min =   -36  Ave =    -4.82  Std =     8.97
   Min TM value:-36 at an offset of 126.450 usec -> 2015-10-13T18:12:52.334.607.200 (XO: 10.3347489)
   Max TM value:28 at an offset of 134.613 usec -> 2015-10-13T18:12:52.334.615.362 (XO: 10.3347570)
   Max dTM/dt [1/us]::4.02e+02 at an offset of 127.850 usec -> 2015-10-13T18:12:52.334.608.600 (XO: 10.3347503)
   Min dTM/dt [1/us]::-3.84e+02 at an offset of 126.838 usec -> 2015-10-13T18:12:52.334.607.587 (XO: 10.3347493)

 - Block number: 2 starting at 2015-10-13T18:13:10.163.938.750 (XO: 28.1640806, t=35.6°C)  Rank: 00034
   TM values: Max =    31  Min =   -45  Ave =    -4.04  Std =     9.52
   Min TM value:-45 at an offset of 3.063 usec -> 2015-10-13T18:13:10.163.941.812 (XO: 28.1640836)
   Max TM value:31 at an offset of 33.813 usec -> 2015-10-13T18:13:10.163.972.562 (XO: 28.1641144)
   Max dTM/dt [1/us]::3.12e+02 at an offset of 130.300 usec -> 2015-10-13T18:13:10.164.069.050 (XO: 28.1642109)
   Min dTM/dt [1/us]::-3.22e+02 at an offset of 131.713 usec -> 2015-10-13T18:13:10.164.070.462 (XO: 28.1642123)

 - Block number: 3 starting at 2015-10-13T18:13:13.278.187.750 (XO: 31.2783294, t=35.6°C)  Rank: 00007
   TM values: Max =    37  Min =   -45  Ave =    -4.77  Std =    12.26
   Min TM value:-45 at an offset of 94.050 usec -> 2015-10-13T18:13:13.278.281.800 (XO: 31.2784234)
   Max TM value:37 at an offset of 119.400 usec -> 2015-10-13T18:13:13.278.307.150 (XO: 31.2784488)
   Max dTM/dt [1/us]::4.48e+02 at an offset of 119.063 usec -> 2015-10-13T18:13:13.278.306.812 (XO: 31.2784484)
   Min dTM/dt [1/us]::-4.00e+02 at an offset of 131.350 usec -> 2015-10-13T18:13:13.278.319.100 (XO: 31.2784607)
