-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151013_105715_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.911073 to 31.913653
-each separately from 1s to 31s and from 449991us to 913653us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.538629 in 1 packets - #/No: 18/18
      16.463084 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-13T10:57:19.911.073.000 to 2015-10-13T10:57:49.913.653.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.538629 in 1 packets - #/No: 18/18
      16.463084 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.3564942 to 32.3564114
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999614 in 1 packets - #/No: 18/18
      15.9999558 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151013_105715_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.2675662 to 3.2678552
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.5385910  -- OK for SL --
   MIU time from 2015-10-13T10:57:19.911.073.000 to 2015-10-13T10:57:19.911.362.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.538629

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8064462 to 16.8067352
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4630390  -- OK for SL --
   MIU time from 2015-10-13T10:57:33.449.991.000 to 2015-10-13T10:57:33.450.280.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.463084

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.2697742 to 33.2700632
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-13T10:57:49.913.364.000 to 2015-10-13T10:57:49.913.653.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-13T10:57:19.910.932.750
   Max SL corrected packet time: 2015-10-13T10:57:49.913.512.750
   Min SL corrected XO rel time: 1.9110745
   Max SL corrected XO rel time: 31.9136544
   TM values from -16 to 6 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-13T10:57:19.910.932.750 (XO: 1.9110745, t=35.6°C)  Rank: 00003
   TM values: Max =     6  Min =   -16  Ave =    -4.75  Std =     3.25
   Min TM value:-16 at an offset of 106.275 usec -> 2015-10-13T10:57:19.911.039.025 (XO: 1.9111807)
   Max TM value:6 at an offset of 41.613 usec -> 2015-10-13T10:57:19.910.974.362 (XO: 1.9111161)
   Max dTM/dt [1/us]::7.60e+01 at an offset of 62.450 usec -> 2015-10-13T10:57:19.910.995.200 (XO: 1.9111369)
   Min dTM/dt [1/us]::-7.50e+01 at an offset of 122.575 usec -> 2015-10-13T10:57:19.911.055.325 (XO: 1.9111970)

 - Block number: 2 starting at 2015-10-13T10:57:33.449.850.750 (XO: 15.4499919, t=35.6°C)  Rank: 00002
   TM values: Max =     5  Min =   -13  Ave =    -4.66  Std =     2.33
   Min TM value:-13 at an offset of 7.525 usec -> 2015-10-13T10:57:33.449.858.275 (XO: 15.4499994)
   Max TM value:5 at an offset of 271.863 usec -> 2015-10-13T10:57:33.450.122.613 (XO: 15.4502637)
   Max dTM/dt [1/us]::5.90e+01 at an offset of 100.913 usec -> 2015-10-13T10:57:33.449.951.662 (XO: 15.4500928)
   Min dTM/dt [1/us]::-6.80e+01 at an offset of 194.401 usec -> 2015-10-13T10:57:33.450.045.150 (XO: 15.4501863)

 - Block number: 3 starting at 2015-10-13T10:57:49.913.223.750 (XO: 31.9133654, t=35.6°C)  Rank: 00003
   TM values: Max =     6  Min =   -14  Ave =    -4.76  Std =     2.56
   Min TM value:-14 at an offset of 48.150 usec -> 2015-10-13T10:57:49.913.271.900 (XO: 31.9134135)
   Max TM value:6 at an offset of 22.950 usec -> 2015-10-13T10:57:49.913.246.700 (XO: 31.9133883)
   Max dTM/dt [1/us]::7.60e+01 at an offset of 22.900 usec -> 2015-10-13T10:57:49.913.246.650 (XO: 31.9133883)
   Min dTM/dt [1/us]::-7.70e+01 at an offset of 210.651 usec -> 2015-10-13T10:57:49.913.434.400 (XO: 31.9135760)
