-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151009_014355_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.197951 to 43.940368
-each separately from 3s to 43s and from 197951us to 940368us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.340353 in 1 packets - #/No: 18/18
      28.401197 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-09T01:44:01.197.951.000 to 2015-10-09T01:44:41.940.368.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.340353 in 1 packets - #/No: 18/18
      28.401197 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4699544 to 44.4698444
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999670 in 1 packets - #/No: 18/18
      27.9999230 in 1 packets - #/No: 36/18
Temperature FPGA: 
      08 -> 44.6°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BE -> 32.4°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      BF -> 34.0°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151009_014355_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.6679058 to 4.6681948
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3403190  -- OK for SL --
   MIU time from 2015-10-09T01:44:01.197.951.000 to 2015-10-09T01:44:01.198.240.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.340353

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0085138 to 17.0088028
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.4011190  -- OK for SL --
   MIU time from 2015-10-09T01:44:13.538.593.000 to 2015-10-09T01:44:13.538.882.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.401197

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.4099218 to 45.4102108
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-09T01:44:41.940.079.000 to 2015-10-09T01:44:41.940.368.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-09T01:44:01.197.810.750
   Max SL corrected packet time: 2015-10-09T01:44:41.940.227.750
   Min SL corrected XO rel time: 3.1979519
   Max SL corrected XO rel time: 43.9403690
   TM values from -17 to 8 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-09T01:44:01.197.810.750 (XO: 3.1979519, t=32.4°C)  Rank: 00004
   TM values: Max =     5  Min =   -12  Ave =    -4.73  Std =     2.18
   Min TM value:-12 at an offset of 95.313 usec -> 2015-10-09T01:44:01.197.906.062 (XO: 3.1980472)
   Max TM value:5 at an offset of 70.550 usec -> 2015-10-09T01:44:01.197.881.300 (XO: 3.1980225)
   Max dTM/dt [1/us]::7.70e+01 at an offset of 304.701 usec -> 2015-10-09T01:44:01.198.115.450 (XO: 3.1982566)
   Min dTM/dt [1/us]::-7.50e+01 at an offset of 68.725 usec -> 2015-10-09T01:44:01.197.879.475 (XO: 3.1980207)

 - Block number: 2 starting at 2015-10-09T01:44:13.538.452.750 (XO: 15.5385939, t=32.4°C)  Rank: 00004
   TM values: Max =     4  Min =   -14  Ave =    -4.74  Std =     2.12
   Min TM value:-14 at an offset of 31.350 usec -> 2015-10-09T01:44:13.538.484.100 (XO: 15.5386252)
   Max TM value:4 at an offset of 34.975 usec -> 2015-10-09T01:44:13.538.487.725 (XO: 15.5386288)
   Max dTM/dt [1/us]::7.30e+01 at an offset of 50.013 usec -> 2015-10-09T01:44:13.538.502.762 (XO: 15.5386439)
   Min dTM/dt [1/us]::-8.30e+01 at an offset of 51.075 usec -> 2015-10-09T01:44:13.538.503.825 (XO: 15.5386449)

 - Block number: 3 starting at 2015-10-09T01:44:41.939.938.750 (XO: 43.9400800, t=32.4°C)  Rank: 00008
   TM values: Max =     8  Min =   -17  Ave =    -4.65  Std =     2.73
   Min TM value:-17 at an offset of 92.638 usec -> 2015-10-09T01:44:41.940.031.387 (XO: 43.9401727)
   Max TM value:8 at an offset of 75.450 usec -> 2015-10-09T01:44:41.940.014.200 (XO: 43.9401555)
   Max dTM/dt [1/us]::9.70e+01 at an offset of 242.338 usec -> 2015-10-09T01:44:41.940.181.088 (XO: 43.9403224)
   Min dTM/dt [1/us]::-1.12e+02 at an offset of 282.438 usec -> 2015-10-09T01:44:41.940.221.188 (XO: 43.9403625)
