-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151006_060005_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.696068 to 38.289417
-each separately from 13s to 38s and from 289128us to 772476us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.075830 in 1 packets - #/No: 18/18
      16.516652 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-06T06:00:21.696.068.000 to 2015-10-06T06:00:46.289.417.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.075830 in 1 packets - #/No: 18/18
      16.516652 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4497311 to 39.4496626
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999781 in 1 packets - #/No: 18/18
      16.9999534 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151006_060005_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.1457988 to 15.1460878
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0758070  -- OK for SL --
   MIU time from 2015-10-06T06:00:21.696.068.000 to 2015-10-06T06:00:21.696.357.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.075830

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2218948 to 23.2221838
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5166070  -- OK for SL --
   MIU time from 2015-10-06T06:00:29.772.187.000 to 2015-10-06T06:00:29.772.476.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.516652

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      25 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.7387908 to 39.7390798
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-06T06:00:46.289.128.000 to 2015-10-06T06:00:46.289.417.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-06T06:00:21.695.927.750
   Max SL corrected packet time: 2015-10-06T06:00:46.289.276.750
   Min SL corrected XO rel time: 13.6960696
   Max SL corrected XO rel time: 38.2894180
   TM values from -21 to 14 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-06T06:00:21.695.927.750 (XO: 13.6960696, t=34.0°C)  Rank: 00005
   TM values: Max =    11  Min =   -21  Ave =    -4.77  Std =     4.72
   Min TM value:-21 at an offset of 22.913 usec -> 2015-10-06T06:00:21.695.950.662 (XO: 13.6960926)
   Max TM value:11 at an offset of 23.788 usec -> 2015-10-06T06:00:21.695.951.537 (XO: 13.6960934)
   Max dTM/dt [1/us]::1.75e+02 at an offset of 85.613 usec -> 2015-10-06T06:00:21.696.013.362 (XO: 13.6961553)
   Min dTM/dt [1/us]::-1.73e+02 at an offset of 20.713 usec -> 2015-10-06T06:00:21.695.948.462 (XO: 13.6960904)

 - Block number: 2 starting at 2015-10-06T06:00:29.772.046.750 (XO: 21.7721878, t=34.0°C)  Rank: 00003
   TM values: Max =     8  Min =   -16  Ave =    -4.72  Std =     3.24
   Min TM value:-16 at an offset of 81.350 usec -> 2015-10-06T06:00:29.772.128.100 (XO: 21.7722691)
   Max TM value:8 at an offset of 269.401 usec -> 2015-10-06T06:00:29.772.316.150 (XO: 21.7724572)
   Max dTM/dt [1/us]::1.34e+02 at an offset of 268.001 usec -> 2015-10-06T06:00:29.772.314.750 (XO: 21.7724558)
   Min dTM/dt [1/us]::-1.53e+02 at an offset of 208.938 usec -> 2015-10-06T06:00:29.772.255.688 (XO: 21.7723967)

 - Block number: 3 starting at 2015-10-06T06:00:46.288.987.750 (XO: 38.2891290, t=34.0°C)  Rank: 00025
   TM values: Max =    14  Min =   -20  Ave =    -4.00  Std =     4.88
   Min TM value:-20 at an offset of 293.476 usec -> 2015-10-06T06:00:46.289.281.225 (XO: 38.2894225)
   Max TM value:14 at an offset of 11.663 usec -> 2015-10-06T06:00:46.288.999.412 (XO: 38.2891406)
   Max dTM/dt [1/us]::1.84e+02 at an offset of 134.138 usec -> 2015-10-06T06:00:46.289.121.887 (XO: 38.2892631)
   Min dTM/dt [1/us]::-1.96e+02 at an offset of 151.925 usec -> 2015-10-06T06:00:46.289.139.675 (XO: 38.2892809)
