-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151005_204017_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.650490 to 30.453678
-each separately from 11s to 30s and from 403140us to 650779us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.752361 in 1 packets - #/No: 18/18
      15.049960 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-05T20:40:31.650.490.000 to 2015-10-05T20:40:50.453.678.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.752361 in 1 packets - #/No: 18/18
      15.049960 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.4663718 to 31.4663196
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.9999890 in 1 packets - #/No: 18/18
      14.9999588 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151005_204017_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.1168608 to 13.1171498
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.7523510  -- OK for SL --
   MIU time from 2015-10-05T20:40:31.650.490.000 to 2015-10-05T20:40:31.650.779.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.752361

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8695008 to 16.8697898
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0499190  -- OK for SL --
   MIU time from 2015-10-05T20:40:35.403.140.000 to 2015-10-05T20:40:35.403.429.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.049960

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9197088 to 31.9199978
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-05T20:40:50.453.389.000 to 2015-10-05T20:40:50.453.678.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-05T20:40:31.650.349.750
   Max SL corrected packet time: 2015-10-05T20:40:50.453.537.750
   Min SL corrected XO rel time: 11.6504908
   Max SL corrected XO rel time: 30.4536794
   TM values from -32 to 23 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-05T20:40:31.650.349.750 (XO: 11.6504908, t=34.0°C)  Rank: 00006
   TM values: Max =    21  Min =   -31  Ave =    -4.74  Std =     8.63
   Min TM value:-31 at an offset of 139.488 usec -> 2015-10-05T20:40:31.650.489.237 (XO: 11.6506302)
   Max TM value:21 at an offset of 248.438 usec -> 2015-10-05T20:40:31.650.598.188 (XO: 11.6507392)
   Max dTM/dt [1/us]::4.00e+02 at an offset of 139.525 usec -> 2015-10-05T20:40:31.650.489.275 (XO: 11.6506303)
   Min dTM/dt [1/us]::-4.04e+02 at an offset of 140.938 usec -> 2015-10-05T20:40:31.650.490.687 (XO: 11.6506317)

 - Block number: 2 starting at 2015-10-05T20:40:35.402.999.750 (XO: 15.4031411, t=34.0°C)  Rank: 00005
   TM values: Max =    23  Min =   -32  Ave =    -4.66  Std =     7.94
   Min TM value:-32 at an offset of 133.775 usec -> 2015-10-05T20:40:35.403.133.525 (XO: 15.4032749)
   Max TM value:23 at an offset of 109.863 usec -> 2015-10-05T20:40:35.403.109.612 (XO: 15.4032510)
   Max dTM/dt [1/us]::4.86e+02 at an offset of 133.813 usec -> 2015-10-05T20:40:35.403.133.562 (XO: 15.4032749)
   Min dTM/dt [1/us]::-4.08e+02 at an offset of 109.900 usec -> 2015-10-05T20:40:35.403.109.650 (XO: 15.4032510)

 - Block number: 3 starting at 2015-10-05T20:40:50.453.248.750 (XO: 30.4533904, t=34.0°C)  Rank: 00004
   TM values: Max =    18  Min =   -30  Ave =    -4.82  Std =     7.13
   Min TM value:-30 at an offset of 110.688 usec -> 2015-10-05T20:40:50.453.359.437 (XO: 30.4535011)
   Max TM value:18 at an offset of 108.588 usec -> 2015-10-05T20:40:50.453.357.337 (XO: 30.4534990)
   Max dTM/dt [1/us]::4.07e+02 at an offset of 110.725 usec -> 2015-10-05T20:40:50.453.359.475 (XO: 30.4535011)
   Min dTM/dt [1/us]::-3.75e+02 at an offset of 109.963 usec -> 2015-10-05T20:40:50.453.358.712 (XO: 30.4535004)
