-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151001_110832_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.358203 to 44.881171
-each separately from 1s to 44s and from 358203us to 881171us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.019998 in 1 packets - #/No: 18/18
      20.502103 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-01T11:08:36.358.203.000 to 2015-10-01T11:09:19.881.171.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.019998 in 1 packets - #/No: 18/18
      20.502103 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.3638299 to 45.3637129
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.9999374 in 1 packets - #/No: 18/18
      19.9999456 in 1 packets - #/No: 36/18
Temperature FPGA: 
      08 -> 44.6°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151001_110832_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      20 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.7220334 to 2.7223224
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.0199350  -- OK for SL --
   MIU time from 2015-10-01T11:08:36.358.203.000 to 2015-10-01T11:08:36.358.492.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.019998

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      21 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.7422574 to 25.7425464
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5020470  -- OK for SL --
   MIU time from 2015-10-01T11:08:59.378.490.000 to 2015-10-01T11:08:59.378.779.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.502103

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.2445934 to 46.2448824
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-01T11:09:19.880.882.000 to 2015-10-01T11:09:19.881.171.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-01T11:08:36.358.062.750
   Max SL corrected packet time: 2015-10-01T11:09:19.881.030.750
   Min SL corrected XO rel time: 1.3582045
   Max SL corrected XO rel time: 44.8811710
   TM values from -40 to 91 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-01T11:08:36.358.062.750 (XO: 1.3582045, t=34.0°C)  Rank: 00020
   TM values: Max =    62  Min =   -39  Ave =    -4.71  Std =     6.62
   Min TM value:-39 at an offset of 77.638 usec -> 2015-10-01T11:08:36.358.140.387 (XO: 1.3582821)
   Max TM value:62 at an offset of 72.850 usec -> 2015-10-01T11:08:36.358.135.600 (XO: 1.3582773)
   Max dTM/dt [1/us]::1.23e+02 at an offset of 151.163 usec -> 2015-10-01T11:08:36.358.213.912 (XO: 1.3583556)
   Min dTM/dt [1/us]::-1.48e+02 at an offset of 52.725 usec -> 2015-10-01T11:08:36.358.115.475 (XO: 1.3582572)

 - Block number: 2 starting at 2015-10-01T11:08:59.378.349.750 (XO: 24.3784911, t=34.0°C)  Rank: 00021
   TM values: Max =    91  Min =   -40  Ave =    -4.74  Std =     6.68
   Min TM value:-40 at an offset of 107.600 usec -> 2015-10-01T11:08:59.378.457.350 (XO: 24.3785987)
   Max TM value:91 at an offset of 105.800 usec -> 2015-10-01T11:08:59.378.455.550 (XO: 24.3785969)
   Max dTM/dt [1/us]::2.88e+02 at an offset of 105.675 usec -> 2015-10-01T11:08:59.378.455.425 (XO: 24.3785968)
   Min dTM/dt [1/us]::-2.12e+02 at an offset of 106.238 usec -> 2015-10-01T11:08:59.378.455.987 (XO: 24.3785974)

 - Block number: 3 starting at 2015-10-01T11:09:19.880.741.750 (XO: 44.8808820, t=34.0°C)  Rank: 00015
   TM values: Max =    40  Min =   -25  Ave =    -4.60  Std =     5.72
   Min TM value:-25 at an offset of 123.550 usec -> 2015-10-01T11:09:19.880.865.300 (XO: 44.8810055)
   Max TM value:40 at an offset of 117.713 usec -> 2015-10-01T11:09:19.880.859.462 (XO: 44.8809997)
   Max dTM/dt [1/us]::1.19e+02 at an offset of 142.088 usec -> 2015-10-01T11:09:19.880.883.837 (XO: 44.8810241)
   Min dTM/dt [1/us]::-1.07e+02 at an offset of 4.613 usec -> 2015-10-01T11:09:19.880.746.362 (XO: 44.8808866)
