-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20150930_111753_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.075713 to 35.742909
-each separately from 2s to 35s and from 75713us to 799946us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.723655 in 1 packets - #/No: 18/18
      18.942674 in 1 packets - #/No: 36/18
MIU packet time from 2015-09-30T11:17:58.075.713.000 to 2015-09-30T11:18:31.742.909.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.723655 in 1 packets - #/No: 18/18
      18.942674 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7306481 to 36.7305588
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999621 in 1 packets - #/No: 18/18
      18.9999486 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20150930_111753_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.8063617 to 3.8066507
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7236150  -- OK for SL --
   MIU time from 2015-09-30T11:17:58.075.713.000 to 2015-09-30T11:17:58.076.002.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.723655

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.5302657 to 18.5305547
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.9426230  -- OK for SL --
   MIU time from 2015-09-30T11:18:12.799.657.000 to 2015-09-30T11:18:12.799.946.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.942674

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.4731777 to 37.4734667
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-09-30T11:18:31.742.620.000 to 2015-09-30T11:18:31.742.909.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-09-30T11:17:58.075.572.750
   Max SL corrected packet time: 2015-09-30T11:18:31.742.768.750
   Min SL corrected XO rel time: 2.0757138
   Max SL corrected XO rel time: 35.7429098
   TM values from -20 to 13 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-09-30T11:17:58.075.572.750 (XO: 2.0757138, t=34.0°C)  Rank: 00006
   TM values: Max =    11  Min =   -20  Ave =    -4.63  Std =     3.48
   Min TM value:-20 at an offset of 52.875 usec -> 2015-09-30T11:17:58.075.625.625 (XO: 2.0757667)
   Max TM value:11 at an offset of 46.175 usec -> 2015-09-30T11:17:58.075.618.925 (XO: 2.0757600)
   Max dTM/dt [1/us]::7.40e+01 at an offset of 107.663 usec -> 2015-09-30T11:17:58.075.680.412 (XO: 2.0758215)
   Min dTM/dt [1/us]::-8.20e+01 at an offset of 9.850 usec -> 2015-09-30T11:17:58.075.582.600 (XO: 2.0757237)

 - Block number: 2 starting at 2015-09-30T11:18:12.799.516.750 (XO: 16.7996577, t=34.0°C)  Rank: 00003
   TM values: Max =    13  Min =   -15  Ave =    -4.73  Std =     2.59
   Min TM value:-15 at an offset of 104.513 usec -> 2015-09-30T11:18:12.799.621.262 (XO: 16.7997622)
   Max TM value:13 at an offset of 98.263 usec -> 2015-09-30T11:18:12.799.615.012 (XO: 16.7997559)
   Max dTM/dt [1/us]::6.50e+01 at an offset of 144.375 usec -> 2015-09-30T11:18:12.799.661.125 (XO: 16.7998020)
   Min dTM/dt [1/us]::-6.70e+01 at an offset of 145.525 usec -> 2015-09-30T11:18:12.799.662.275 (XO: 16.7998032)

 - Block number: 3 starting at 2015-09-30T11:18:31.742.479.750 (XO: 35.7426208, t=34.0°C)  Rank: 00003
   TM values: Max =     6  Min =   -15  Ave =    -4.61  Std =     2.83
   Min TM value:-15 at an offset of 34.625 usec -> 2015-09-30T11:18:31.742.514.375 (XO: 35.7426555)
   Max TM value:6 at an offset of 29.725 usec -> 2015-09-30T11:18:31.742.509.475 (XO: 35.7426506)
   Max dTM/dt [1/us]::7.40e+01 at an offset of 204.126 usec -> 2015-09-30T11:18:31.742.683.875 (XO: 35.7428250)
   Min dTM/dt [1/us]::-8.30e+01 at an offset of 255.551 usec -> 2015-09-30T11:18:31.742.735.300 (XO: 35.7428764)
