-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20150929_232233_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.844477 to 32.344206
-each separately from 2s to 32s and from 343917us to 844766us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.669650 in 1 packets - #/No: 18/18
      10.829212 in 1 packets - #/No: 36/18
MIU packet time from 2015-09-29T23:22:38.844.477.000 to 2015-09-29T23:23:08.344.206.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.669650 in 1 packets - #/No: 18/18
      10.829212 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.5067972 to 33.5067157
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.9999484 in 1 packets - #/No: 18/18
      10.9999701 in 1 packets - #/No: 36/18
Temperature FPGA: 
      08 -> 44.6°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20150929_232233_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3512733 to 4.3515623
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.6695990  -- OK for SL --
   MIU time from 2015-09-29T23:22:38.844.477.000 to 2015-09-29T23:22:38.844.766.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.669650

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.0211613 to 23.0214503
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8291830  -- OK for SL --
   MIU time from 2015-09-29T23:22:57.514.416.000 to 2015-09-29T23:22:57.514.705.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.829212

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.8506333 to 33.8509223
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-09-29T23:23:08.343.917.000 to 2015-09-29T23:23:08.344.206.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-09-29T23:22:38.844.336.750
   Max SL corrected packet time: 2015-09-29T23:23:08.344.065.750
   Min SL corrected XO rel time: 2.8444784
   Max SL corrected XO rel time: 32.3442075
   TM values from -29 to 16 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-09-29T23:22:38.844.336.750 (XO: 2.8444784, t=34.0°C)  Rank: 00004
   TM values: Max =    16  Min =   -25  Ave =    -4.69  Std =     6.18
   Min TM value:-25 at an offset of 158.088 usec -> 2015-09-29T23:22:38.844.494.837 (XO: 2.8446365)
   Max TM value:16 at an offset of 246.051 usec -> 2015-09-29T23:22:38.844.582.800 (XO: 2.8447244)
   Max dTM/dt [1/us]::2.88e+02 at an offset of 136.350 usec -> 2015-09-29T23:22:38.844.473.100 (XO: 2.8446147)
   Min dTM/dt [1/us]::-2.94e+02 at an offset of 144.313 usec -> 2015-09-29T23:22:38.844.481.062 (XO: 2.8446227)

 - Block number: 2 starting at 2015-09-29T23:22:57.514.275.750 (XO: 21.5144171, t=34.0°C)  Rank: 00004
   TM values: Max =    15  Min =   -24  Ave =    -4.60  Std =     6.36
   Min TM value:-24 at an offset of 135.338 usec -> 2015-09-29T23:22:57.514.411.087 (XO: 21.5145524)
   Max TM value:15 at an offset of 155.550 usec -> 2015-09-29T23:22:57.514.431.300 (XO: 21.5145726)
   Max dTM/dt [1/us]::2.77e+02 at an offset of 4.088 usec -> 2015-09-29T23:22:57.514.279.837 (XO: 21.5144212)
   Min dTM/dt [1/us]::-2.80e+02 at an offset of 4.150 usec -> 2015-09-29T23:22:57.514.279.900 (XO: 21.5144212)

 - Block number: 3 starting at 2015-09-29T23:23:08.343.776.750 (XO: 32.3439185, t=34.0°C)  Rank: 00004
   TM values: Max =    16  Min =   -29  Ave =    -4.60  Std =     6.77
   Min TM value:-29 at an offset of 118.525 usec -> 2015-09-29T23:23:08.343.895.275 (XO: 32.3440371)
   Max TM value:16 at an offset of 94.388 usec -> 2015-09-29T23:23:08.343.871.137 (XO: 32.3440129)
   Max dTM/dt [1/us]::3.05e+02 at an offset of 94.888 usec -> 2015-09-29T23:23:08.343.871.637 (XO: 32.3440134)
   Min dTM/dt [1/us]::-2.97e+02 at an offset of 126.113 usec -> 2015-09-29T23:23:08.343.902.862 (XO: 32.3440446)
