-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20150929_030935_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.022400 to 32.188790
-each separately from 1s to 32s and from 22400us to 724367us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.701389 in 1 packets - #/No: 18/18
      2.464134 in 1 packets - #/No: 36/18
MIU packet time from 2015-09-29T03:09:39.022.400.000 to 2015-09-29T03:10:10.188.790.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.701389 in 1 packets - #/No: 18/18
      2.464134 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4144298 to 33.4143457
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      27.9999240 in 1 packets - #/No: 18/18
      2.9999919 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      07 -> 45.9°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      08 -> 44.6°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20150929_030935_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4368310 to 2.4371200
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.7013110  -- OK for SL --
   MIU time from 2015-09-29T03:09:39.022.400.000 to 2015-09-29T03:09:39.022.689.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.701389

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      33 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.1384310 to 31.1387200
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.4641270  -- OK for SL --
   MIU time from 2015-09-29T03:10:07.724.078.000 to 2015-09-29T03:10:07.724.367.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.464134

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.6028470 to 33.6031360
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-09-29T03:10:10.188.501.000 to 2015-09-29T03:10:10.188.790.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-09-29T03:09:39.022.259.750
   Max SL corrected packet time: 2015-09-29T03:10:10.188.649.750
   Min SL corrected XO rel time: 1.0224013
   Max SL corrected XO rel time: 32.1887908
   TM values from -69 to 102 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-09-29T03:09:39.022.259.750 (XO: 1.0224013, t=34.0°C)  Rank: 00006
   TM values: Max =    12  Min =   -22  Ave =    -4.64  Std =     5.49
   Min TM value:-22 at an offset of 109.800 usec -> 2015-09-29T03:09:39.022.369.550 (XO: 1.0225111)
   Max TM value:12 at an offset of 113.463 usec -> 2015-09-29T03:09:39.022.373.212 (XO: 1.0225147)
   Max dTM/dt [1/us]::2.86e+02 at an offset of 103.725 usec -> 2015-09-29T03:09:39.022.363.475 (XO: 1.0225050)
   Min dTM/dt [1/us]::-2.75e+02 at an offset of 131.550 usec -> 2015-09-29T03:09:39.022.391.300 (XO: 1.0225328)

 - Block number: 2 starting at 2015-09-29T03:10:07.723.937.750 (XO: 29.7240792, t=34.0°C)  Rank: 00033
   TM values: Max =   102  Min =   -69  Ave =    -4.60  Std =    10.62
   Min TM value:-69 at an offset of 71.850 usec -> 2015-09-29T03:10:07.724.009.600 (XO: 29.7241510)
   Max TM value:102 at an offset of 69.150 usec -> 2015-09-29T03:10:07.724.006.900 (XO: 29.7241483)
   Max dTM/dt [1/us]::3.20e+02 at an offset of 79.538 usec -> 2015-09-29T03:10:07.724.017.287 (XO: 29.7241587)
   Min dTM/dt [1/us]::-3.63e+02 at an offset of 73.875 usec -> 2015-09-29T03:10:07.724.011.625 (XO: 29.7241530)

 - Block number: 3 starting at 2015-09-29T03:10:10.188.360.750 (XO: 32.1885018, t=34.0°C)  Rank: 00008
   TM values: Max =    27  Min =   -38  Ave =    -4.53  Std =     6.25
   Min TM value:-38 at an offset of 118.250 usec -> 2015-09-29T03:10:10.188.479.000 (XO: 32.1886201)
   Max TM value:27 at an offset of 121.588 usec -> 2015-09-29T03:10:10.188.482.337 (XO: 32.1886234)
   Max dTM/dt [1/us]::2.39e+02 at an offset of 20.775 usec -> 2015-09-29T03:10:10.188.381.525 (XO: 32.1885226)
   Min dTM/dt [1/us]::-2.15e+02 at an offset of 130.125 usec -> 2015-09-29T03:10:10.188.490.875 (XO: 32.1886319)
