-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20150928_123847_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.441031 to 41.399429
-each separately from 6s to 41s and from 37355us to 441320us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.596035 in 1 packets - #/No: 18/18
      21.361496 in 1 packets - #/No: 36/18
MIU packet time from 2015-09-28T12:38:56.441.031.000 to 2015-09-28T12:39:31.399.429.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.596035 in 1 packets - #/No: 18/18
      21.361496 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4336578 to 42.4335639
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999625 in 1 packets - #/No: 18/18
      20.9999436 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20150928_123847_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.8746883 to 7.8749773
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.5959990  -- OK for SL --
   MIU time from 2015-09-28T12:38:56.441.031.000 to 2015-09-28T12:38:56.441.320.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.596035

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.4709763 to 21.4712653
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3614390  -- OK for SL --
   MIU time from 2015-09-28T12:39:10.037.355.000 to 2015-09-28T12:39:10.037.644.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.361496

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.8327043 to 42.8329933
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-09-28T12:39:31.399.140.000 to 2015-09-28T12:39:31.399.429.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-09-28T12:38:56.440.890.750
   Max SL corrected packet time: 2015-09-28T12:39:31.399.288.750
   Min SL corrected XO rel time: 6.4410316
   Max SL corrected XO rel time: 41.3994304
   TM values from -17 to 12 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-09-28T12:38:56.440.890.750 (XO: 6.4410316, t=34.0°C)  Rank: 00006
   TM values: Max =     7  Min =   -16  Ave =    -4.75  Std =     2.98
   Min TM value:-16 at an offset of 52.538 usec -> 2015-09-28T12:38:56.440.943.287 (XO: 6.4410842)
   Max TM value:7 at an offset of 89.800 usec -> 2015-09-28T12:38:56.440.980.550 (XO: 6.4411214)
   Max dTM/dt [1/us]::5.60e+01 at an offset of 163.913 usec -> 2015-09-28T12:38:56.441.054.662 (XO: 6.4411956)
   Min dTM/dt [1/us]::-5.80e+01 at an offset of 82.713 usec -> 2015-09-28T12:38:56.440.973.462 (XO: 6.4411144)

 - Block number: 2 starting at 2015-09-28T12:39:10.037.214.750 (XO: 20.0373561, t=34.0°C)  Rank: 00005
   TM values: Max =    12  Min =   -17  Ave =    -4.72  Std =     2.98
   Min TM value:-17 at an offset of 39.738 usec -> 2015-09-28T12:39:10.037.254.487 (XO: 20.0373958)
   Max TM value:12 at an offset of 34.300 usec -> 2015-09-28T12:39:10.037.249.050 (XO: 20.0373904)
   Max dTM/dt [1/us]::6.80e+01 at an offset of 8.800 usec -> 2015-09-28T12:39:10.037.223.550 (XO: 20.0373649)
   Min dTM/dt [1/us]::-7.40e+01 at an offset of 35.013 usec -> 2015-09-28T12:39:10.037.249.762 (XO: 20.0373911)

 - Block number: 3 starting at 2015-09-28T12:39:31.398.999.750 (XO: 41.3991414, t=34.0°C)  Rank: 00006
   TM values: Max =     6  Min =   -15  Ave =    -4.63  Std =     2.95
   Min TM value:-15 at an offset of 54.125 usec -> 2015-09-28T12:39:31.399.053.875 (XO: 41.3991956)
   Max TM value:6 at an offset of 152.125 usec -> 2015-09-28T12:39:31.399.151.875 (XO: 41.3992936)
   Max dTM/dt [1/us]::7.10e+01 at an offset of 98.513 usec -> 2015-09-28T12:39:31.399.098.262 (XO: 41.3992400)
   Min dTM/dt [1/us]::-6.20e+01 at an offset of 30.800 usec -> 2015-09-28T12:39:31.399.030.550 (XO: 41.3991722)
